The paper deals with the implementation of the BBM method which enables to measure direct Auger spectraN(E) into the Auger spectrometer with energy modulation on CMA. The arrangement of the modificated Auger spectrometer on the instrumental, hardware and software level is presented. Moreover, the system includes the optional function — the analog background separation — which enables the optimalization of weak Auger signal measurements in a higher energy region. The implementation of the BBM method is convenient to use as a means of innovation of older Auger spectrometers which enable to get the differentiated spectra only.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Seah H. P.: Surface Interf. Analysis1 (1979) 86.
Langeron J. P. and Minel L.: Surface Science138 (1984) 61.
EP Groupe: Jeol News E20 (1983) No. 3.
Chattarji D.: The Theory of Auger Transitions. Academic Press, London, 1976.
High Resolution Auger Electron Spectrometer System, Instruction Manual fy Varian No. 87-400 338.
Tomek S. and Harman R.: Report III-7-2/03. Electrotekh. faculty, Slovak Technical University, Bratislava, 1990 (in Slovak).
Tomek S., Liday J., and Vogrinčič P.: Čs. čas. fyz.42 (1992) 273 (in Slovak).
About this article
Cite this article
Tomek, S., Liday, J. & Vogrinčič, P. The innovation facilities of Auger spectrometer with energy modulation on CMA. Czech J Phys 44, 287–293 (1994). https://doi.org/10.1007/BF01694494
- Energy Region
- Energy Modulation
- Signal Measurement
- Optional Function