Numerical analysis of the flow characteristics of rotary blood pump
- 213 Downloads
Thrombus formation and hemolysis have been linked to the dynamics of blood flow in rotary blood pumps and ventricular assist devices. Hemolysis occurs as the blood passes through the pump housing, and thrombi develop in stagnation and low-velocity regions. The predicted velocities, pressure, and turbulence quantities from the numerical simulation are used to identify regions of high shear stress and internal recirculation. A nimerical technique is described that simulates the hydrodynamic characteristics of a rotary blood pump with a flow rate of 6 l/min at a rotational speed of 3000 RPM. A computational fluid dynamics (CFD) code, CFX 4, is used to solve the time-dependent incompressible Navier-Stokes equations using a transient finite volume method and three-dimensional structured grids. The simulation utilized the sliding mesh capabilities of this numerical code to model the rotating impeller and examine the effect of blade shape on the hydrodynamic performance of the blood pump in terms of pressure rise, flow rates, and energy losses. The first impeller model has six straight channels; the second impeller has six backward-curved channels. The results for two impeller configurations are presented and discussed. The curvedpump design resulted in higher pressure rise and maximum shear stresses than the straight-channel one. In general the paper demonstrates that CFD is an essential numerical tool for optimizing pump performance with the aim of reducing trauma to the blood cells.
Key wordsRotary blood pump Numerical simulation Hydrodynamics Centrifugal blood pump CFD
Unable to display preview. Download preview PDF.
- 1.Hellums GD, Brown CH. Blood cell damage by mechanical forces. In: Wang V, Normann NA (eds) Cardiovascular flow dynamics and measurement. Baltimore: University Park Press, 1977;799–823Google Scholar
- 2.Golding LB, Groves LK, Peters M, Jacobs G, Sukalac R, Nose Y, Loop FD. Initial clinical experience with a new temporary left venticular assist device. Ann Throrac Surg 1980;29:66–69Google Scholar
- 3.Baldwin JT, Deutsch S, Geselowitz DB, Tarbell JM. Estimation of Reynolds stresses within the Penn State left ventricular assist device. Trans ASAIO 1990;36:M274-M278Google Scholar
- 7.Takiura K, Masuzawa T, Endo S, Wakisaka Y, Tatsumu E, Taenaka Y, Takano H, Yamane T, Nishida M, Asztalos B, Konishi Y, Miyazoe Y, Ito K. Development of design methods of a centrifugal blood pump with in vitro tests, flow visualisation and computational fluid dynamics: results in hemolysis tests. Artif Organs 1998;22:393–398PubMedGoogle Scholar
- 8.Schima H, Huber L, Glekas JP, Siegl H, Muller MR, Wieselthaler G, Losert U, Thoma H, Wolner E. Three approaches for flow evaluation in centrifugal blood pumps: numerical calculation, visualization and in-vitro thrombus formation (abstract). Artif Organs 1993;17:471Google Scholar
- 10.Pinotti M, Panone N, Tomasini EP. Laser Doppler velocity measurements in a centrifugal ventricular assist device. In: Durao DFG, Heitor MV (eds) Proceedings of the 7th International Symposium on Applications of Laser Techniques to Fluid Mechanics. Lisbon, Portugal: Instituto superior Tecnico 1994;321–325Google Scholar
- 12.Nishida H, Yamaki F, Naktani H, Endo M, Koyanagi H, Oshiyama H, Horiuchi K, Kijima T, Nojiri C, Fukasawa H, Akutsu T. Development of the Terumo Capiox centrifugal pump and its clinical application to open heart surgery: a comparative study with the roller pump. Artif Organs 1993;17:323–327PubMedGoogle Scholar
- 15.AEA Technology, CFX 4.1 Flow Solver User Guide, Harwell Laboratory, UK, 1995Google Scholar
- 16.Morsi YS, Holand PG, Clayton BR. Prediction of turbulent swirling flows in axisymmetric annuli. J Appl Math Model 1995;19:M613-M620Google Scholar