An acoustical particle counter (or acoustical particle sizing device) was evaluated for counting and sizing of fog droplets. Fog droplets were generated under controlled conditions in the laboratory. Fog droplet sizes were measured by an optical and an electron microscope and compared to results from the acoustical particle counter. Most of the droplets were found to be in the size range of 5–30 µm. The mean diameters estimated from the acoustical particle counter were in agreement with the microscope values of mean droplet diameter. A Rich 100 condensation nuclei monitor was also operated simultaneously during the fog droplet counting to monitor the condensation nuclei counts in the laboratory. The results indicate that condensation nuclei count is inversely correlated with the fog droplet threshold diameter. Aerosols of uniform size (35 µm) were generated by the vibrating orifice monodisperse aerosol generator and counted at three different flow rates by the acoustical particle counter. The counts/liter/minute were similar, indicating the reliability of the acoustical particle counter.
This is a preview of subscription content, log in to check access.
Buy single article
Instant access to the full article PDF.
Price includes VAT for USA
Subscribe to journal
Immediate online access to all issues from 2019. Subscription will auto renew annually.
This is the net price. Taxes to be calculated in checkout.
Chemical Corps: 1952–1955, ‘Chemical Corps Biological Laboratory Contract DA-18-064-CML-2296’,Quarterly Reports.
Langer, G.: 1965, ‘An Acoustical Particle Counter-Pereliminary Results’,J. Colloid. Sci. 20, 602–609.
Langer, G.: 1969, ‘Status of Acoustic Particle Counter Research’,Powder Technology 2, 307–309.
Langer, G.: 1973, ‘Evaluation of NCAR Ice Nucleus Counter, Part I: Basic Operation’,J. App. Meteorol. 12, 1000–1011.
Langer, G., Cooper, G., Nagamoto, C.T. and Rosinski, J.: 1978, ‘Ice Nucleation Mechanisms of Submicron Monodispersed Silver Iodide, 1, 5-Dihydroxynapthalene and Phloroglucinol Aerosol Particles’,J. App. Meteorol. 17, 1039–1048.
Cadle, R.D., Shapiro M.A. and Langer, G.: 1979, ‘Concentrations of Condensation Nuclei in the Vicinity of Jetstream Maxima’,Arch. Meteorol. Geophys. Bioclimatol. Ser. A. 28, 1–10.
Coover, S.R.: 1978, ‘Acoustical Particle Sizing Device’, Ph.D. Dissertation, Department of Environmental Sciences and Engineering, University of North Carolina, Chapel Hill, NC.
Reist, P.C. and Burgess, W.A.: 1968, ‘A Comparative Evaluation of Three Aerosol Methods’,Am. Ind. Hyg. Assoc. J. 29, 123–128.
Coover, S.R. and Reist, P.C.: 1980, ‘Acoustical Particle Sizing Device’,Environ. Sci. Technol. 14, 951–954.
Instruction Manual for Model 3050 Berglund-Liu Vibrating Orifice Monodisperse Aerosol Generator.
Reist, P.C.: 1984,Introduction to Aerosol Science, Macmillan Publishing Company, New York, Ch. 12, pp. 181–203.
Batton, L.J. and Reitan, C.H.: 1957,Artificial Stimulation of Rain, Pergamon Press, New York, p. 184.
About this article
Cite this article
Singh, D.P., Reist, P.C. Evaluation of acoustical particle counter for the sizing of fog droplets. Environ Monit Assess 24, 35–44 (1993). https://doi.org/10.1007/BF00568798
- Particle Size
- Electron Microscope
- Environmental Management
- Size Range
- Droplet Size