# Spatial resolution and measurement of turbulence in the viscous sublayer using subminiature hot-wire probes

- Received:

- 156 Citations
- 654 Downloads

## Abstract

Measurements in the viscous sublayer of a flat-plate turbulent boundary layer in air, using single hot-wire sensors with lengths from 1–60 viscous length scales show that, at a given distance from the surface, the turbulence intensity, flatness factor, and skewness factor of the longitudinal velocity fluctuation are nearly independent of wire length when the latter is less than 20–25 times the viscous length scale (i.e. 20–25 “wall units”), and decrease significantly and abruptly for larger wire lengths. This conclusion is consistent with other workers' probability density functions of streak spacing: the lateral spacing of “streaks” in the viscous sublayer is 80–100 wall units on average with minimum spacing of 20–25 wall units, which implies that signals would be strongly attenuated by wires whose length exceeds 20–25 wall units. To achieve wire lengths of less than 20–25 wall units, subminiature hot wire probes like those described by Ligrani and Bradshaw (1987), having lengths as small as 150 μm, are necessary for sublayer measurements in typical laboratory wind tunnels. As well as the measurements mentioned above, dissipation spectra are presented, to show the effect of spanwise averaging on the high-frequency motions, which is necessarily more severe than the effect on overall intensities.

## Preview

Unable to display preview. Download preview PDF.

### References

- Blackwelder, R. F.; Haritonidis, J. H. 1983: Scaling of the bursting frequency in turbulent boundary layers. J. Fluid Mech. 132, 87–103Google Scholar
- Blackwelder, R. F.; Kaplan, R. E. 1976: On the wall-structure of the turbulent boundary layer. J. Fluid Mech. 76, 89–112Google Scholar
- Bradshaw, P. 1967: The turbulence structure of equilibrium boundary layers. J. Fluid Mech. 29, 625–645Google Scholar
- Bradshaw, P. 1972: Two-more wind tunnels driven by aerofoiltype centrifugal blowers. I.C. Aero. Report 72-10, April 1972, Imperial College, London, GBGoogle Scholar
- Bremhorst, K. 1972: The effect of wire length and separation on X-array hot-wire anemometer measurements. IEEE Trans. Instrum. Meas. 21, 244–248Google Scholar
- Collis, D. C.; Williams, M. J. 1959: Two-dimensional convection from heated wires at low Reynolds numbers. J. Fluid Mech. 6, 357–384Google Scholar
- Corrsin, S.; Kovasznay, L. S. G. 1949: On the hot-wire length correction. Phys. Rev. 75, 1954Google Scholar
- Derksen, R. W.; Azad R. S. 1983: An examination of hot-wire length corrections. Phys. Fluids 26, 1751–1754Google Scholar
- Dryden, H. L.; Schubauer, G. B.; Mock, W. C.; Skramstad, H. K 1937: Measurements of intensity and scale of wind-tunnel turbulence and their relation to the critical Reynolds number of spheres. NACA Technical Report 581Google Scholar
- Eckelmann, H. 1974: The structure of the viscous sublayer and the adjacent region in a turbulent channel flow. J. Fluid Mech. 65, 439–459Google Scholar
- Frenkiel, F. N. 1949: The influence of the length of a hot wire on the measurements of turbulence. Phys. Rev. 75, 1263–1264Google Scholar
- Frenkiel, F. N. 1954: Effects of wire length in turbulence investigations with a hot-wire anemometer. Aeronaut. Q. 5, 1–24Google Scholar
- Hancock, P. E. 1980: The effect of free-stream turbulence on turbulent boundary layers. Ph.D. Thesis, Department of Aeronautics, Imperial College, London, GBGoogle Scholar
- Hinze, J. O. 1975: Turbulence. (2nd ed.) New York: McGraw-HillGoogle Scholar
- Hussain, A. K M. F.; Reynolds, W. C. 1975: Measurements in fully developed'turbulent channel flow. ASME Trans. J. Fluids Eng. 97, 568–578Google Scholar
- Johansson, A. V.; Alfredsson, P. H. 1982: On the structure of turbulent channel flow. J. Fluid Mech. 122, 295–314Google Scholar
- Johansson, A. V.; Alfredsson, P. H. 1983: Effects of imperfect spatial resolution on measurements of wall-bounded turbulent shear flows. J. Fluid Mech. 135, 409–421Google Scholar
- Karlsson, R. 1980: Studies of skin friction in turbulent boundary layers on smooth and rough walls. Doctoral Thesis, Dept. of Applied Thermo and Fluid Dynamics, Chalmers Univ. Tesch., Gøteborg, SwedenGoogle Scholar
- Kastrinakis, L.; Wallace, J. M.; Willmarth, W. W. 1975: Measurements of small scale streamwise vorticity in a turbulent channel flow. Bull. Am. Phys. Soc. 20, 1422Google Scholar
- Kim, H. T.; Kline, S. J.; Reynolds, W. C. 1971: The production of turbulence near a smooth wall in a turbulent boundary layer. J. Fluid Mech. 50, 133–160Google Scholar
- Klebanoff, P. S. 1954: Characteristics of turbulence in a boundary layer with zero pressure gradient. NACA Technical Note 3178Google Scholar
- Kline, S. J.; Reynolds, W. C.; Schraub, F. A.; Runstalder, P. W. 1967: The structure of turbulent boundary layers. J. Fluid Mech. 30, 741–773Google Scholar
- Kreplin, H.-P.; Eckelmann, H. 1979: Behavior of the three fluctuating velocity components in the wall region of a turbulent channel flow. Phys. Fluids 22, 1233–1239Google Scholar
- Lee, M. K.; Eckelman, L. D.; Hanratty, T. J. 1974: Identification of turbulent wall eddies through the phase relation of the components of the fluctuating velocity gradient. J. Fluid Mech. 66, 17–33Google Scholar
- Ligrani, P. M. 1984: Subminiature hot-wire sensor construction. Report NPS69-84-010, Department of Mechanical Engineering, Naval Postgraduate School, Monterey/CA, USAGoogle Scholar
- Ligrani, P. M.; Bradshaw, P. 1987: Subminiature hot wire sensors: development and use. J. Phys. E. (in print)Google Scholar
- Ligrani, P. M.; Moffat, R. J. 1986: Structure of transitionally rough and fully rough turbulent boundary layers. J. Fluid Mech. 162, 69–98Google Scholar
- Oldaker, D. K.; Tiederman, W. G. 1977: Spatial structure of the viscous sublayer in drag-reducing channel flows. Phys. Fluids 20, 5133–5144Google Scholar
- Pao, Y. 1965: Structure of turbulent velocity and scalar fields at large wavenumbers. Phys. Fluids 8, 1063–1075Google Scholar
- Purtell, L. P.; Klebanoff, P. S.; Buckley, F. T. 1981: Turbulent boundary layer at low Reynolds number. Phys. Fluids 24, 802–811Google Scholar
- Roberts, J. B. 1973: On the correction of hot wire turbulence measurements for spatial resolution errors. Aeronaut. J. 77, 406–412Google Scholar
- Smith, C. R. 1984: A synthesized model of the near-wall behavior in turbulent boundary layers. In: Proceedings of Eighth Symposium on Turbulence (eds. Patterson, G. K.; Zakin, J. L.). Department of Chemical Engineering, University of Missouri at Rolla, USAGoogle Scholar
- Smith, C. R.; Metzler, S. P. 1983: The characteristics of low-speed streaks in the near-wall region of a turbulent boundary layer. J. Fluid Mech. 129, 27–54Google Scholar
- Smol'yakov, A. V.; Tkachenko, V. M. 1983: The measurement of turbulent fluctuations: An introduction to hot-wire anemometry and related transducers. (ed. Bradshaw, P.). Berlin, Heidelberg, New York: Springer-VerlagGoogle Scholar
- Tennekes, H.; Lumley, J. L. 1972: A first course in turbulence. Boston: MIT PressGoogle Scholar
- Uberoi, M. S.; Kovasznay, L. S. G. 1953: On mapping and measurement of randon fields. Q. Appl. Math. 10, 375–393Google Scholar
- Willmarth, W. W.; Bogar, T. J. 1977: Survey and new measurements of turbulent structures near the wall. Phys. Fluids 20, S9-S21Google Scholar
- Willmarth, W. W.; Sharma, L. K. 1984: Study of turbulent structure with hot wires smaller then the viscous length. J. Fluid Mech. 142, 121–149Google Scholar
- Wyngaard, J. C. 1968: Measurement of small-scale turbulence structure with hot wires. J. Phys. E. 1, 1105–1108Google Scholar
- Wyngaard, J. C. 1969: Spatial resolution of the vorticity meter and other hot-wire arrays. J. Phys. E. 2, 983–987Google Scholar
- Wyngaard, J. C. 1971: Spatial resolution of a resistance wire temperatur sensor. Phys. Fluids 14, 2052–2054Google Scholar