A system for measuring leaf gas exchange based on regulating vapour pressure difference
- 120 Downloads
A system for measurement of leaf gas exchange while regulating leaf to air vapour pressure difference has been developed; it comprises an assimilation chamber, leaf temperature controller, mass flow controller, dew point controller and personal computer. A relative humidity sensor and air and leaf temperature sensors, which are all used for regulating the vapour pressure difference, are mounted into the chamber. During the experiments, the computer continuously monitored the photosynthetic parameters and measurement conditions, so that accurate and intenstive measurements could be made.
When measuring the light-response curve of CO2 assimilation for single leaves, in order to regulate the vapour pressure difference, the leaf temperature and relative humidity in the chamber were separately and simultaneously controlled by changing the air temperature around the leaf and varying the air flow rate through the chamber, respectively. When the vapour pressure difference was regulated, net CO2 assimilation, transpiration and leaf conductance for leaves of rice plant increased at high quantum flux density as compared with those values obtained when it was not regulated.
When measuring the temperature-response curve of CO2 assimilation, the regulation of vapour pressure difference was manipulated by the feed-forward control of the dew point temperature in the inlet air stream. As the vapour pressure difference was regulated at 12 mbar, the maximum rate of and the optimum temperature for CO2 assimilation in rice leaves increased 5 μmolCO2 m−2 s−1 and 5°C, respectively, as compared with those values obtained when the vapour pressure difference took its own course. This was reasoned to be due to the increase in leaf conductance and the decrease in transpiration rate. In addition, these results confirmed that stomatal conductance essentially increases with increasing leaf temperature under constant vapour pressure difference conditions, in other words, when the influence of the vapour pressure difference is removed.
This system may be used successfully to measure inter- and intra-specific differences and characteristics of leaf gas exchange in plants with a high degree of accuracy.
Key wordscontrol measurement photosynthesis vapour pressure difference
CO2 assimilation rate
Maximum rate of CO2 assimilation
Optimum teperature for CO2 assimilation
Controlled-temperature water bath
Dew point controller
Transpiration rate; gl, leaf conductance
Humidity control circuit
Infrared gas analyzer
Leaf temperature controller
Mass flow controller
Quantum flux density
Relative humidity controller
Vapour pressure difference
Difference of CO2 concentration between inlet and outlet air
Unable to display preview. Download preview PDF.
- 1.Gaastra P (1959) Photosynthesis of crop plants as influenced by light, carbon dioxide, temperature and stomatal diffusion resistance. Meded LandHoogesch, Wageningen 59: 1–68Google Scholar
- 2.Hall AE and Kaufman MR (1975) Stomatal response to environment with Sesamum indicum L. Plant Physiol 55: 455–59Google Scholar
- 3.Hall AE, Camacho-B SE and Kaufman MR (1975) Regulation of water loss by citrus leaves. Physiol Plant 33: 62–65Google Scholar
- 4.Jarvis PG and Mansfield TA (1981) Stomatal Physiology. Cambridge University Press, CambridgeGoogle Scholar
- 5.Kanda A, Imai K, Moriya T, Handa S and Terajima T (1984) A gas exchange measurement system and its application to rice and cassava leaves. Japan Jour Crop Sci 53: 472–478Google Scholar
- 6.Koch W, Lange OL and Schulze E-D (1971) Ecophysiological investigations on wild and cultivated plant in the negev desert. I. Methods: A mobile laboratory for measuring carbon dioxide and water vapour exchange. Oecologia 8: 296–309Google Scholar
- 7.Lösch R (1979) Responses of stomata to environmental factors. Experiments with isolated epidermal strips of Polypodium vulgare. II. Leaf bulk water potential, air humidity, and temperature. Oecologia 39: 229–238Google Scholar
- 8.Schulze E-D (1972) A new type of climatized gas exchange chamber for net photosynthesis and transpiration measurements in the field. Oecologia 10: 243–251Google Scholar
- 9.Schulze E-D and Hall AE (1982) Stomatal responses, water loss and CO2 assimilation rates of plants in contrasting environments. In: Lange OL et al. eds. Physiological plant ecology II, pp 181–230. Encyclopedia of plant physiology. New Series, Volume 12B, Springer, BerlinGoogle Scholar
- 10.Sestak Z, Čatsky J and Jarvis PG (eds) (1971) Plant photosynthetic production: Manual of methods. The Hague: Dr W Junk PublishersGoogle Scholar
- 11.Turner NC, Schulze E-D and Gollan T (1984) The responses of stomata and leaf gas exchange to vapour pressure deficits and soil water content. I. Species comparisons at high soil water contents. Oecologia 63: 338–342Google Scholar
- 12.von Caemmerer S and Farquhar GD (1981) Some relationships between the biochemistry of photosynthesis and the gas exchange of leaves. Planta 153: 376–387Google Scholar