Calibration of Watermark soil moisture sensors for soil matric potential and temperature
- 450 Downloads
Rapid, accurate, and automated measurement of soil matric potential is desirable. Evidence suggested that the Watermark resistance block might be an appropriate and inexpensive tool, so we conducted an evaluation of its relevant characteristics. A number of these blocks were calibrated under laboratory conditions to determine their individual and aggregate responses to soil matric potential, soil type, and temperature. We found that the temperature response could be expressed as a single equation, valid for all tested blocks, but comparison against matric potential revealed that each block had a characteristic response. Furthermore, block responses were different in two soils and, for a given soil, not necessarily reproducible. Given these limitations, these sensors are probably useful only as relative indicators of soil water status.
Key wordscalibration moisture block resistance block soil matric potential temperature
Unable to display preview. Download preview PDF.
- Armstrong C F, Ligon J T and Thomson S J 1985 Calibration of Watermark Model 200 soil moisture sensor. ASAE Paper No. 85–2077. ASAE, St. Joseph, MI 49085.Google Scholar
- Bouyoucos G J and Mick A H 1940 An electrical resistance method for the continuous measurement of soil moisture under field conditions. Michigan Agric. Exp. Sta. Tech. Bull. 172.Google Scholar
- Campbell G S and Gee G W 1986 Water potential: Miscellaneous methods. In Methods of Soil Analysis, Part 1, 2nd ed. Ed. AKlute. pp 619–633. American Society of Agronomy, Madison, WI.Google Scholar
- Hillel D 1980 Fundamentals of Soil Physics. Academic Press, San Diego, CA. 403 p.Google Scholar