Bench and mathematical modeling of the effects of breathing a helium/oxygen mixture on expiratory time constants in the presence of heterogeneous airway obstructions
Expiratory time constants are used to quantify emptying of the lung as a whole, and emptying of individual lung compartments. Breathing low-density helium/oxygen mixtures may modify regional time constants so as to redistribute ventilation, potentially reducing gas trapping and hyperinflation for patients with obstructive lung disease. In the present work, bench and mathematical models of the lung were used to study the influence of heterogeneous patterns of obstruction on compartmental and whole-lung time constants.
A two-compartment mechanical test lung was used with the resistance in one compartment held constant, and a series of increasing resistances placed in the opposite compartment. Measurements were made over a range of lung compliances during ventilation with air or with a 78/22% mixture of helium/oxygen. The resistance imposed by the breathing circuit was assessed for both gases. Experimental results were compared with predictions of a mathematical model applied to the test lung and breathing circuit. In addition, compartmental and whole-lung time constants were compared with those reported by the ventilator.
Time constants were greater for larger minute ventilation, and were reduced by substituting helium/oxygen in place of air. Notably, where time constants were long due to high lung compliance (i.e. low elasticity), helium/oxygen improved expiratory flow even for a low level of resistance representative of healthy, adult airways. In such circumstances, the resistance imposed by the external breathing circuit was significant. Mathematical predictions were in agreement with experimental results. Time constants reported by the ventilator were well-correlated with those determined for the whole-lung and for the low-resistance compartment, but poorly correlated with time constants determined for the high-resistance compartment.
It was concluded that breathing a low-density gas mixture, such as helium/oxygen, can improve expiratory flow from an obstructed lung compartment, but that such improvements will not necessarily affect time constants measured by the ventilator. Further research is required to determine if alternative measurements made at the ventilator level are predictive of regional changes in ventilation. It is anticipated that such efforts will be aided by continued development of mathematical models to include pertinent physiological and pathophysiological phenomena that are difficult to reproduce in mechanical test systems.
- Farré R, Navajas D: Assessment of expiratory flow limitation in chronic obstructive pulmonary disease: a new approach. Eur Respir J 2004, 23:187–188. CrossRef
- O'Donnell DE: Hyperinflation, dyspnea, and exercise intolerance in chronic obstructive pulmonary disease. Proc Am Thoracic Soc 2006, 3:180–184. CrossRef
- Permutt S: The role of the large airways on smooth muscle contraction in asthma. J Appl Physiol 2007, 103:1457–1458. CrossRef
- Barbini P, Brighenti C, Gnudi G: A simulation study of expiratory flow limitation in obstructive patients during mechanical ventilation. Annals of Biomed Eng 2006, 34:1879–1889. CrossRef
- Brighenti C, Barbini P, Gnudi G, Cevenini G, Pecchiari M, D'Angelo E: Helium-oxygen ventilation in the presence of expiratory flow-limitation: A model study. Respir Physiol Neurobiol 2007, 157:326–334. CrossRef
- D'Angelo E, Santus P, Civitillo MF, Centanni S, Pecchiari M: Expiratory flow-limitation and heliox breathing in resting and exercising COPD patients. Respir Physiol Neurobiol 2009, 169:291–296. CrossRef
- Pecchiari M, Pelucchi A, D'Angelo E, Foresi A, Milic-Emili J, D'Angelo E: Effect of heliox breathing on dynamic hyperinflation in COPD patients. Chest 2004, 125:2075–2082. CrossRef
- Palange P, Valli G, Onorati P, Antonucci R, Paoletti P, Rosato A, Manfredi F, Serra P: Effect of heliox on lung dynamic hyperinflation, dyspnea, and exercise endurance capacity in COPD patients. J Appl Physiol 2004, 97:1637–1642. CrossRef
- Manthous CA, Hall JB, Melmed A, Caputo MA, Walter J, Klocksieben JM, Schmidt GA, Wood LDH: Heliox improves pulsus paradoxus and peak expiratory flow in nonintubated patients with severe asthma. Am J Respir Crit Care Med 1995, 151:310–314.
- Papamoschou D: Theoretical validation of the respiratory benefits of helium-oxygen mixtures. Respir Physiol 1995, 99:183–190. CrossRef
- Katz IM, Martin AR, Muller P-A, Terzibachi K, Feng C-H, Caillibotte G, Sandeau J, Texereau J: The ventilation distribution of helium-oxygen mixtures and the role of inertial losses in the presence of heterogeneous airway obstructions. J Biomechanics 2011, 44:1137–1143. CrossRef
- de Lange EE, Altes TA, Patrie JT, Battiston JJ, Juersvich AP, Mugler JP III, Platts-Mills TA: Changes in regional airflow obstruction over time in the lungs of patients with asthma: Evaluation with 3He MR imaging. Radiology 2009, 250:567–575. CrossRef
- Venegas JG, Schroeder T, Harris S, Winkler RT, Melo MFV: The distribution of ventilation during bronchoconstriction is patchy and bimodal: A PET imaging study. Resp Physiol Neurobiol 2005, 148:57–64. CrossRef
- Kurahashi K, Ohtsuka M, Usuda Y: Central airway occlusion underestimates intrinsic positive end-expiratory pressure: a numerical and physical simulation. Exp Lung Res 2009, 35:756–769. CrossRef
- Diehl J-L, Peigne V, Guérot E, Faisy C, Lecourt L, Mercat A: Helium in the adult critical care setting. Annals of Intensive Care 2011, 1:24. CrossRef
- Otis AB, McKerrow CB, Bartlett RA, Mead J, McIlroy MB, Selverstone NJ, Radford EP: Mechanical factors in distribution of pulmonary ventilation. J Appl Physiol 1956, 8:427–443.
- Silva CA M, Ventura CE: A simple model illustrating the respiratory system’s time constant concept. Adv Physiol Educ 2006, 30:129–130. CrossRef
- Brunner JX, Laubscher TP, Banner MJ, Iotti G, Braschi A: Simple method to measure total expiratory time constant based on the passive expiratory flow-volume curve. Crit Care Med 1995, 23:1117–1122. CrossRef
- Despas PJ, Leroux M, Macklem PT: Site of airway obstruction in asthma as determined by measuring maximal expiratory flow breathing air and a helium-oxygen mixture. J Clin Investig 1972, 51:3235–3243. CrossRef
- Hamilton Medical: Hamilton G5 Operator's Manual. 2008, 6–22. 624074/02
- Katz IM, Martin AR, Feng C-H, Majoral C, Caillibotte G, Marx T, Bazin J-E, Daviet C: Airway pressure distribution during xenon anesthesia: the insufflation phase at constant flow (volume controlled mode). Appl Cardiopulm Pathophysiol 2012, 16:5–16.
- Lichtwarck-Aschoff M, Helmer A, Kawati R, Lattuada M, Sjostrand UH, Zugel N, Guttman J, Hedenstierna G: Good short-term agreement between measured and calculated tracheal pressure. Brit J Anesth 2003, 91:239–48. CrossRef
- Jarreau P-H, Louis B, Dassieu G, Desfrere L, Blanchard PW, Moriette G, Isabey D, Harf A: Estimation of inspiratory pressure drop in neonatal and pediatric endotracheal tubes. J Appl Physiol 1999, 87:36–46.
- Katz I, Caillibotte G, Martin AR, Arpentinier P: Property value estimation for inhaled therapeutic binary gas mixtures: He, Xe, N2O, and N2 with O2. Med Gas Res 2011, 1:28. CrossRef
- Bench and mathematical modeling of the effects of breathing a helium/oxygen mixture on expiratory time constants in the presence of heterogeneous airway obstructions
- Open Access
- Available under Open Access This content is freely available online to anyone, anywhere at any time.
BioMedical Engineering OnLine
- Online Date
- May 2012
- Online ISSN
- BioMed Central
- Additional Links
- Time constant
- Gas trapping
- Ventilation distribution
- Airway resistance
- Mechanical ventilation
- Author Affiliations
- 1. Senior Research Scientist, Medical Gases, American Air Liquide, Delaware Research and Technology Center (DRTC), 200 GBC Drive, Newark, DE, 19702, USA
- 2. Medical Gases Group, Air Liquide Santé International, Les Loges-en-Josas, 78354, France
- 3. Department of Mechanical Engineering, Lafayette College, Easton, PA, 18042, USA