Classification Performance of a Low Pressure Differential Mobility Analyzer for Nanometer-sized Particles
Purchase on Springer.com
$39.95 / €34.95 / £29.95*
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.
Size distribution measurements and classification tests by a low pressure differential mobility analyzer (LPDMA) for nanometer-sized silver particles and cesium iodide particles under low pressure conditions (123–300 Torr) were performed using a transmission electron microscope (TEM) and the tandem DMA technique. When the ratio of the sheath gas flow rate and the aerosol gas flow rate was set at 5 : 1, the targeted sizes calculated from the classification voltage applied to the LPDMA at 160 Torr are found to be in good correlation with the count mean Feret diameter obtained from the TEM observation of the particles collected after a classification ranging from 6 to 25 nm, although the targeted sizes set by the LPDMA were approximately 15% greater than the count mean Feret diameter measured by the TEM after classification. The geometric standard deviations of DMA-classified particles measured by TEM and those obtained from the tandem DMA method ranged from 1.08 to 1.17 and from 1.05 to 1.13, respectively. They were slightly greater than the ideal geometric standard deviations (1.05) of the particles classified with the LPDMA, which was calculated by neglecting the broadening effects due to Brownian diffusion. We experimentally demonstrated the validity of our LPDMA system for size measurements and classification of the nanometer-sized particles under low pressure conditions.
- Camata R.P., H.A. Atwater, K.J. Vahala & R.C. Flagan, 1996. Size classification of silicon nanocrystals. Appl. Phys. Lett. 68, 3162–3164.
- Deppert K., J.O. Bovin, J.O. Malm & L. Samuelson, 1996. A new method to fabricate size-selected compound semiconductor nanocrystals: aerotaxy, J. Crystal Growth 169, 13–19.
- Hummes D., S. Neumann, F. Schmidt, M. Dröboom, H. Fissan, K. Deppert, T. Junno, J.O. Malm & L. Samuelson, 1996. Determination of the size distribution of nanometer-sized particles. J. Aerosol Sci. 27S, 163–164.
- Knutson E.O. & K.T. Whitby, 1975. Aerosol classification by electric mobility: apparatus, theory, and applications. J. Aerosol Sci. 6, 443–451.
- Kousaka Y., K. Okuyama & M. Adachi, 1985a. Determination of particle size distribution of ultra-fine aerosols using a differential mobility analyzer. Aerosol Sci. Technol. 4, 209–225.
- Kousaka Y., K. Okuyama, M. Adachi & T. Miura, 1985b. Effect of Brownian diffusion on electrical classification of ultrafine aerosol particles in differential mobility analyzer. J. Chem. Eng. Japan 5, 401–407.
- Kousaka Y., Y. Endo, H. Ichitsubo & M. Alonso, 1996. Orientation-specific dynamic shape factors for doublets and triplets of spheres in the transition regime. Aerosol Sci. Technol. 24, 36–44.
- Kuga Y., S. Endoh, S. Satooka, Y. Okada, T. Tashiro & K. Takeuchi, 1991. Collection of 235U-enriched UF5 particles and direct measurement of separation factor in laser isotope separation of uranium. J. Nucl. Sci. Technol. 28, 870–872.
- Kuga Y., B.J. Jurcik, S. Satooka & K. Takeuchi, 1995. Improvement of multi-jet low pressure impactor for high collection efficiency of UF5 in the molecular laser isotope separation of uranium. J. Nucl. Sci. Matter. 224, 43–49.
- Kuga Y., S. Isomura, K. Takeuchi & K. Okuyama, 1996. Growth enhancement of UF5 nano-particles assisted by α-ray ionization. Appl. Phys. A 62, 373–379.
- Kuga, Y., M. Hirasawa, T. Seto, K. Okuyama & K. Takeuchi, 1999. In situ observation of UF5 nanoparticle growth in a lowpressure mixed-flow reactor. Appl. Phys. A 68, 75–80.
- Kuga Y., Y. Furuyama, K. Ando, K. Okuyama & K. Takeuchi, 2000. Nanometer-sized silver particle measurements by low pressure differential mobility analyzer and its classification performance (In Japanese). Kagaku Kougaku Ronbunshu 26, 776–784.
- Rader D.J. & P.H. McMurry, 1983. Application of the tandem differential mobility analyzer to studies of droplet growth or evaporation. J. Aerosol Sci. 17, 771–787.
- Rosell-Llompart J. & J. Fernandez de la Mora, 1993. Minimization of the diffusive broadening of ultrafine particles in differential mobility analyzers. In Marijnissen J. & Pratsinis S., eds. Synthesis and Characterization of Ultra fine Particles, Delft Univ. Press, Delft.
- Rosell-Llompart J., I.G. Loscertales, D. Bingham & J. Fernandez de la Mora, 1996. Sizing nanoparticles and ions with a short differential mobility analyzer. J. Aerosol Sci. 27, 695–719.
- Scheibel H.G. & J. Porstendörfer, 1983. Generation of monodisperse Ag-and NaCl-aerosols with particle diameters between 2 and 300 nm. J. Aerosol Sci. 14, 113–126.
- Seto T., T. Nakamoto, K. Okuyama, M. Adachi, Y. Kuga & K. Takeuchi, 1997. Size distribution measurement of nanometer-sized aerosol particles using DMA under lowpressure conditions. J. Aerosol Sci. 28, 193–206.
- Takeuchi K., Y. Kuga, S. Satooka, & K. Okuyama, 1998. Growth of UF5 nanoparticles formed by laser photolysis in a supersonic nozzle. J. Aerosol Sci. 29, 1027–1033.
- Winklmayr W., G.P. Reischl, A.O. Linder & A. Berner, 1991. A new electromobility spectrometer for the measurement of aerosol size distribution in the size range from 1 to 1000 nm. J. Aerosol Sci. 22, 289–296.
- Classification Performance of a Low Pressure Differential Mobility Analyzer for Nanometer-sized Particles
Journal of Nanoparticle Research
Volume 3, Issue 2-3 , pp 175-183
- Cover Date
- Print ISSN
- Online ISSN
- Kluwer Academic Publishers
- Additional Links
- differential mobility analyzer (DMA)
- low pressure
- tandem DMA
- Industry Sectors