An Evaporation-Based Disposable Micropump Concept for Continuous Monitoring Applications
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.Get Access
An inexpensive and simple pumping principle is described that is capable of delivering both small and constant flow rates (10–1,000 nl/min) over a longer period of time (days to weeks). The concept is based on controlled evaporation of a liquid through a membrane into a gas space containing a sorption agent. As long as the sorption agent keeps the vapor pressure in the gas phase below saturation, fluid evaporated from the membrane is replaced by capillary forces inducing flow from a reservoir. In a feasibility study, a total volume of 300 μl of Ringer's solution has been continuously pumped over a period of six days, resulting in an constant average flow rate of 35 nl/min (590 pl/s). The maximum liquid volume transported is limited by sorption capacity and amount of the sorption agent. Low fabrication costs, high reliability (no moving parts), the suitability for integration into planar system architectures and the lack of a special external energy source besides an environment of regulated temperature are important features of the concept, in particular with regard to its potential application in continuous patient monitoring. Truly continuous flow can be achieved in contrast to many other pump mechanisms leading to discontinuous, pulse-type flow. A challenge for a broader range of applications is the inherent temperature dependence of the flow rate. In its current version, the pump can only be used in a suction-mode.
- H. Andersson, W. Vander Wijngaart, P. Nilsson, P. Enoksson, and G. Stemme, Sensors & Actuators B 72, 259-265 (2001).
- P. Bergveld, Biomedical Microdevices 2, 185-195 (2000).
- S. Böhm, W. Olthuis, and P. Bergveld, Biomedical Microdevices 1, 121-130 (2000).
- T.A. Desai, D.J. Hansford, L. Kulinsky, A.H. Nashat, G. Rasi, J. Tu, Y. Wang, M. Ahang, and M. Ferrari, Biomedical Microdevices 2, 11-40 (1999).
- C.S. Effenhauser, Topics in Current Chemistry 194, 51-82 (1998).
- M. Elwenspoek, T.S.J. Lammerink, R. Miyake, and J.H.J. Fluitman, Journal of Micromechanics Micro-engineering 4, 227-245 (1994).
- L.M. Hullegie, H.L. Lutgers, R.P.F. Dullaart, W.J. Sluiter, K.J. Wientjes, A.J.M. Schoonen, and K. Hoogenberg, The Netherlands Journal of Medicine 57, 13-19 (2000).
- W.A. Kaptein, J.C. Zwaagstra, K. Venema, and J. Korf, Analytical Chemistry 70, 4696-4700 (1998).
- J.P. Kutter, Trends in Analytical Chemistry 19, 352-363 (2000).
- H.L. Lutgers, L.M. Hullegie, K. Hoogenberg, W.J. Sluiter, R.P.F. Dullaart, K.J. Wientjes, and A.J.M. Schoonen, The Netherlands Journal of Medicine 57, 7-12 (2000).
- N.-T. Nguyen and X. Huang, Sensors & Actuators A 88, 104-111 (2001).
- J. Perdomo, H. Hinkers, C. Sundermeier, W. Seifert, O. Martinez Morell, and M. Knoll, Biosensors & Bioelectronics 15, 515-522 (2000).
- R.F. Probstein, “Physicochemical Hydrodynamics”, 2nd edn, Wiley, New York, 1994.
- S. Shoji, Topics in Current Chemistry 194, 163-188 (1998).
- F. Sternberg, C. Meyerhoff, F. Mennel, F. Bischoff, and E.F. Pfeiffer, Diabetes Care 18, 1266-1269 (1995).
- L. Taiz and E. Zeiger, “Plant Physiology”, 2nd edn, Whitaker, 1998.
- T. Tamura, K. Koseki, T. Sumino, M. Ogawa, T. Togawa, and K. Tsuchiya, Frontiers in Medical and Biological Engineering 10, 147-156 (2000).
- M.A. Unger, H.-P. Chou, T. Thorsen, A. Scherer, and S.R. Quake, Science 288, 113-116 (2000).
- T. Vering, S. Adam, H. Drewer, C. Dumschat, R. Steinkuhl, A. Schulee, E.M. Siegel, and M. Knoll, Analyst 123, 1605-1609 (1998).
- A. Wego and L. Pagel, Sensors & Actuators A 88, 220-226 (2001).
- An Evaporation-Based Disposable Micropump Concept for Continuous Monitoring Applications
Volume 4, Issue 1 , pp 27-32
- Cover Date
- Print ISSN
- Online ISSN
- Kluwer Academic Publishers
- Additional Links
- continuous monitoring
- Industry Sectors