Early Out-of-Hospital Experience with an Impedance-Compensating Low-Energy Biphasic Waveform Automatic External Defibrillator
- Roger D. White
- … show all 1 hide
Purchase on Springer.com
$39.95 / €34.95 / £29.95*
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.
Impedance-compensating low-energy biphasic truncated exponential(BTE) waveforms are effective in transthoracic defibrillation ofshort-duration ventricular fibrillation (VF). However, the BTE waveform hasnot been examined in out-of-hospital cardiac arrest (OHCA) with patients inprolonged VF often associated with myocardial ischemia. The objective ofthis study was to evaluate the BTE waveform automatic externaldefibrillator (AED) in the out-of-hospital setting with long-duration VF.AEDs incorporating a 150-J BTE waveform were placed in 12 police squad carsand 4 paramedic-staffed advanced life support ambulances. AEDs were appliedto arrested patients by first-arriving personnel, whether police orparamedics. Data were obtained from PC Data Cards within the AED.Defibrillation was defined as at least transient termination of VF. Tenpatients, 64 ± 14 years, were treated for VF with BTE shocks.Another 8 patients were in nonshockable rhythms and the AEDs,appropriately, did not advise a shock. Five of the 10 VF arrests werewitnessed with a 911 call-to-shock time of 6.6 ± 1.7 minutes. VFdetection and defibrillation occurred in all 10 patients. Spontaneouscirculation was restored in 3 of 5 witnessed arrest patients and 1 survivedto discharge home. Fifty-one VF episodes were converted with 62 shocks. Presenting VF amplitude and rate were 0.43 ± 0.22 (0.13-0.86) mV and232 ± 62 (122-353) beats/min, respectively, and defibrillation wasachieved with the first shock in 7 of 10 patients. Including transientconversions, defibrillation occurred in 42 of 51 VF episodes (82%)with one BTE shock. Shock impedance was 85 ± 10 (39-138) ohms.Delivered energy and peak voltage were 152 ± 2 J and 1754 ±4 V, respectively. The average number of shocks per VF episode was 1.2± 0.5 (1-3). More than one shock was needed in only 9 episodes; nonerequired >3 shocks to defibrillate. Impedance-compensating low-energyBTE waveforms terminated VF in OHCA patients with a conversion rateexceeding that of higher energy monophasic waveforms. VF was terminated inall patients, including those with high impedance.
- Ryan TJ, Anderson JL, Antman, et al. ACC/AHA guidelines for the management of patients with acute myocardial infarction: Executive summary. Circulation 1996;94: 2341–2350.
- Bayes de Luna A, Coumel P, Leclercq JF. Ambulatory sudden cardiac death: Mechanisms of production of fatal arrhythmia on the basis of data from 157 cases. Am Heart J 1989;117:151–159.
- Emergency Cardiac Care Committee and Subcommittees, American Heart Association. Guidelines for cardiopulmonary resuscitation and emergency cardiac care, IX: Ensuring effectiveness of community-wide emergency cardiac care. JAMA 1992;268:2289–2295.
- Weaver WD, Copass MK, Bufi D, et al. Improved neurologic recovery and survival after early defibrillation. Circulation 1984;69:943–948.
- Spaite DW, Hanlon T, Criss EA, et al. Prehospital cardiac arrest: The impact of witnessed collapse and bystander CPR in a metropolitan EMS system with short response times. Ann Emerg Med 1990;19:1264–1269.
- Larsen MP, Eisenberg MS, Cummins RO, et al. Predicting survival from out-of-hospital cardiac arrest: A graphic model. Ann Emerg Med 1993;22:652–1658.
- Weaver WD, Cobb LA, Hallstrom AP, et al. Factors influencing survival after out-of-hospital cardiac arrest. J Am Coll Cardiol 1986;7:752–757.
- Herlitz J, Ekstrom L, Wennerblom B, et al. Survival in patients found to have ventricular fibrillation after cardiac arrest witnessed outside hospital. Eur Heart J 1994;15: 1628–1633.
- Weisfeldt ML, Kerber RE, McGoldrick RP, et al. American Heart Association Report on the Public Access Defibrillation Conference, December 8–10, 1994. Circulation 1995;92: 2740–2747.
- Becker LB, Ostrander MP, Barrett J, et al. Outcome of CPR in a large metropolitan area—Where are the survivors? Ann Emerg Med 1991;20:355–361.
- Lombardi G, Gallagher EJ, Gennis P. Outcome of out-of-hospital cardiac arrest in New York City: The Pre-Hospital Arrest Survival Evaluation (PHASE) study. JAMA 1994; 271:678–683.
- Eisenberg MS, Horwood BT, Cummins RO, et al. Cardiac arrest and resuscitation: A tale of 20 cities. Ann Emerg Med 1990;19:170–186.
- White RD, Vukov LF, Bugliosi TF. Early defibrillation by police: Initial experience with measurement of critical time intervals and patient outcome. Ann Emerg Med 1994; 23:1099–1013.
- White RD, Asplin BR, Bugliosi TF, Hankins DG. High discharge survival rate after out-of-hospital ventricular fibrillation with rapid defibrillation by police and paramedics. Ann Emerg Med 1996;28:480–485.
- Winkle RA, Mead RH, Ruder MA, et al. Improved low energy defibrillation efficacy in man with the use of a biphasic truncated exponential waveform. Am Heart J 1989;117: 122–127.
- Bardy GH, Ivey TD, Allen MD, et al. A prospective, randomized evaluation of biphasic vs. monophasic waveform pulses on defibrillation efficacy in humans. J Am Coll Cardiol 1989;14:728–733.
- Swartz JF, Fletcher RD, Karasik PE. Optimization of biphasic waveforms for human nonthoracotomy defibrillation. Circulation 1993;33:2646–2654.
- Jones JL, Jones RE. Decreased defibrillator-induced dysfunction with biphasic rectangular waveforms. Am J Physiol 1984;247:H792–H796.
- Jones JL, Jones RE, Balasky G. Improved defibrillator waveform safety factor with biphasic waveforms. Am J Physiol 1983;245:H60–H65.
- Osswald S, Trouton TG, O’Nunain SS, et al. Relation between shock-related myocardial injury and defibrillation efficacy of monophasic and biphasic shocks in a canine model. Circulation 1994;90:2501–2509.
- Gurvich NL, Makarychev VA. Defibrillation of the heart with biphasic electrical impulses. Kardiologiya 1967;7: 109–112.
- Kouwenhoven WB. The development of the defibrillator. Ann Intern Med 1969;71:449–457.
- Kerber RE, Becker LB, Bourland JD, et al. Automatic external defibrillators for public access defibrillation: Recommendations for specifying and reporting arrhythmia analysis algorithm performance, incorporating new waveforms, and enhancing safety. A statement for health care professionals from the American Heart Association Task Force on Automatic External Defibrillation, Subcommittee on AED Safety and Efficacy. Circulation 1997;95:1677–1682.
- Bardy GH, Marchlinski FE, Sharma AD, et al. Multicenter comparison of truncated biphasic shocks and standard damped sine wave monophasic shocks for transthoracic ventricular defibrillation. Circulation 1996;94:2507–2514.
- Bardy GH, Gliner BE, Kudenchuk PJ, et al. Truncated biphasic pulses for transthoracic defibrillation. Circulation. 1995;91:1768–1774.
- Echt DS, Barbey JT, Black JN. Influence of ventricular fibrillation duration on defibrillation energy in dogs using bidirectional pulse discharges. PACE 1988;11:1315–1323.
- Jones JL, Swartz JF, Jones RE, Fletcher R. Increasing fibrillation duration enhances relative asymmetrical biphasic versus monophasic defibrillator waveform efficacy. Circ Res 1990;67:376–384.
- Walcot GP, Melnick SB, Chapman FW, et al. Comparison of monophasic and biphasic waveforms for external defibrillation in an animal model of cardiac arrest and resuscitation (abstr) J Am Coll Cardiol 1995;405–406.
- Advanced Cardiac Life Support Textbook. American Heart Association, 1994:4–11.
- Geddes LA, Tacker WA, McFarlane J, Bourland J. Strength-duration curves for ventricular defibrillation in dogs. Circ Res 1970;25:551–560.
- Schuder JC, Stoeckle H, West JA, Keskar PY. Transthoracic ventricular defibrillation in the dog with truncated and untruncated exponential sitmuli. IEEE Trans Biomed Eng 1971;18:1410–415.
- Schuder JC, Rahmoeller GA, Stoekle H. Transthoracic ventricular defibrillation with triangular and trapezoidal waveforms. Circ Res 1996;19:689–694.
- Gliner BE, Lyster TE, Dillon SM, Bardy GH. Transthoracic defibrillation of swine with monophasic and biphasic waveforms. Circulation 1995;92:1634–1643.
- Greene HL, DiMarco JP, Kudenchuk PJ, et al. Comparison of monophasic and biphasic defibrillating pulse waveforms for transthoracic cardioversion. Am J Cardiol 1995;75: 1135–1139.
- Behr JC, Hartley LL, York DK, Brown DD, Kerber RE. Truncated exponential versus damped sinusoidal waveform shocks for transthoracic defibrillation. J Cardiol 1996; 78:1242–1245.
- Weaver WD, Cobb LA, Copass MK, Hallstrom AP. Ventricular defibrillation—a comparative trial using 175-J and 320-J shocks. N Engl J Med 1982;307:1101–1106.
- Reddy RK, Gleva MJ, Dolack GL, et al. Biphasic truncated waveform transthoracic defibrillation results in less postshock ECG ST segment changes than standard damped sine wave shocks (abstr). J Am Coll Cardiol 1995;405.
- Early Out-of-Hospital Experience with an Impedance-Compensating Low-Energy Biphasic Waveform Automatic External Defibrillator
Journal of Interventional Cardiac Electrophysiology
Volume 1, Issue 3 , pp 203-208
- Cover Date
- Print ISSN
- Online ISSN
- Kluwer Academic Publishers
- Additional Links
- sudden death
- automatic external defibrillator
- biphasic waveform
- Industry Sectors
- Roger D. White (1)
- Author Affiliations
- 1. Mayo Clinic and Mayo Medical School; Co-Medical Director, Mayo Clinic Gold Cross Ambulance Service, Inc, Rochester, Minnesota