The 2009 Frank Stinchfield Award: “Hip Squeaking”: A Biomechanical Study of Ceramic-on-ceramic Bearing Surfaces
Purchase on Springer.com
$39.95 / €34.95 / £29.95*
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.
We designed and implemented an in vitro bench test to simulate and identify potential biomechanical causes for hip squeaking with alumina ceramic-on-ceramic bearing surfaces. All bearings were third-generation alumina ceramic with a 32-mm head coupled with a 56-mm acetabular component with a 32-mm ceramic insert. Conditions for testing were normal gait, high load, stripe wear, stripe wear in extreme load, metal transfer, edge wear with extreme load, and microfracture. Each condition was tested two times in dry conditions and two times in a lubricated condition with 25% bovine serum. Squeaking was reproduced in all dry conditions. It occurred quickly with high load, stripe wear, or metal transfer. Once squeaking occurred, it did not stop. Squeaking disappeared for all conditions when a small amount of lubricant was introduced. In lubricated conditions, squeaking was only reproduced for the material transfer condition. Our observations suggest squeaking is a problem of ceramic-ceramic lubrication and that this noise occurs when the film fluid between two surfaces is disrupted. Material (metal) transfer was the only condition that led to squeaking in a lubricated situation.
- Back DL, Dalziel R, Young D, Shimmin A. Early results of primary Birmingham hip resurfacings. An independent prospective study of the first 230 hips. J Bone Joint Surg Br. 2005;87:324–329. CrossRef
- Bierbaum BE, Nairus J, Kuesis D, Morrison JC, Ward D. Ceramic-on-ceramic bearings in total hip arthroplasty. Clin Orthop Relat Res. 2002;405:158–163. CrossRef
- Bohler M, Mochida Y, Bauer TW, Plenk H Jr, Salzer M. Wear debris from two different alumina-on-alumina total hip arthroplasties. J Bone Joint Surg Br. 2000;82:901–909. CrossRef
- Boutin P. Total arthroplasty of the hip by fritted aluminum prosthesis. Experimental study and 1st clinical applications [in French]. Rev Chir Orthop Reparatrice Appar Mot. 1972;58:229–246.
- Boutin P. Total hip arthroplasty using a ceramic prosthesis. Pierre Boutin (1924–1989). Clin Orthop Relat Res. 2000;379:3–11. CrossRef
- Charnley J. Low Friction Arthroplasty of the Hip. Berlin, Germany: Springer-Verlag; 1979.
- Dorlot JM, Christel P, Meunier A. Wear analysis of retrieved alumina heads and sockets of hip prostheses. J Biomed Mater Res. 1989;23(Suppl):299–310. CrossRef
- Hamadouche M, Boutin P, Daussange J, Bolander ME, Sedel L. Alumina-on-alumina total hip arthroplasty: a minimum 18.5-year follow-up study. J Bone Joint Surg Am. 2002;84:69–77.
- Hannouche D, Hamadouche M, Nizard R, Bizot P, Meunier A, Sedel L. Ceramics in total hip replacement. Clin Orthop Relat Res. 2005;430:62–71. CrossRef
- Hannouche D, Nich C, Bizot P, Meunier A, Nizard R, Sedel L. Fractures of ceramic bearings: history and present status. Clin Orthop Relat Res. 2003;417:19–26.
- Jazrawi LM, Bogner E, Della Valle CJ, Chen FS, Pak KI, Stuchin SA, Frankel VH, Di Cesare PE. Wear rates of ceramic-on-ceramic bearing surfaces in total hip implants: a 12-year follow-up study. J Arthroplasty. 1999;14:781–787. CrossRef
- Keurentjes JC, Kuipers RM, Wever DJ, Schreurs BW. High incidence of squeaking in THAs with alumina ceramic-on-ceramic bearings. Clin Orthop Relat Res. 2008;466:1438–1443. CrossRef
- Morlock M, Nassutt R, Janssen R, Willmann G, Honl M. Mismatched wear couple zirconium oxide and aluminum oxide in total hip arthroplasty. J Arthroplasty. 2001;16:1071–1074. CrossRef
- Nevelos J, Ingham E, Doyle C, Streicher R, Nevelos A, Walter W, Fisher J. Microseparation of the centers of alumina-alumina artificial hip joints during simulator testing produces clinically relevant wear rates and patterns. J Arthroplasty. 2000;15:793–795. CrossRef
- Ranawat AS, Ranawat CS. The squeaking hip: a cause for concern—agrees. Orthopedics. 2007;30:738, 743.
- Restrepo C, Parvizi J, Kurtz SM, Sharkey PF, Hozack WJ, Rothman RH. The noisy ceramic hip: is component malpositioning the cause? J Arthroplasty. 2008;23:643–649. CrossRef
- Sedel L. Evolution of alumina-on-alumina implants: a review. Clin Orthop Relat Res. 2000;379:48–54. CrossRef
- Taylor S, Manley MT, Sutton K. The role of stripe wear in causing acoustic emissions from alumina ceramic-on-ceramic bearings. J Arthroplasty. 2007;22(Suppl 3):47–51. CrossRef
- Toni A, Traina F, Stea S, Sudanese A, Visentin M, Bordini B, Squarzoni S. Early diagnosis of ceramic liner fracture. Guidelines based on a twelve-year clinical experience. J Bone Joint Surg Am. 2006;88(Suppl 4):55–63.
- Walter WL, Insley GM, Walter WK, Tuke MA. Edge loading in third generation alumina ceramic-on-ceramic bearings: stripe wear. J Arthroplasty. 2004;19:402–413. CrossRef
- Walter WL, O’Toole GC, Walter WK, Ellis A, Zicat BA. Squeaking in ceramic-on-ceramic hips: the importance of acetabular component orientation. J Arthroplasty. 2007;22:496–503. CrossRef
- Willmann G. Ceramic femoral head retrieval data. Clin Orthop Relat Res. 2000;379:22–28. CrossRef
- Yang CC, Kim RH, Dennis DA. The squeaking hip: a cause for concern—disagrees. Orthopedics. 2007;30:739–742.
- Yoo JJ, Kim YM, Yoon KS, Koo KH, Song WS, Kim HJ. Alumina-on-alumina total hip arthroplasty. A five-year minimum follow-up study. J Bone Joint Surg Am. 2005;87:530–535. CrossRef
- The 2009 Frank Stinchfield Award: “Hip Squeaking”: A Biomechanical Study of Ceramic-on-ceramic Bearing Surfaces
Clinical Orthopaedics and Related Research®
Volume 468, Issue 2 , pp 345-350
- Cover Date
- Print ISSN
- Online ISSN
- Additional Links
- Industry Sectors
- Author Affiliations
- 1. Pavillon T-Département de Chirurgie Orthopédique, Hôpital Edouard Herriot, Lyon, France
- 2. Laboratoire de Biomécanique des Chocs, LBMC–INRETS URMT 9406, Université Claude Bernard, Lyon, France
- 4. Biomechanics Laboratory, Mayo Clinic, Rochester, MN, USA
- 3. Department of Orthopedic Surgery, Mayo Clinic, 200 First Street SW, Rochester, MN, 55905, USA