Hip Resurfacing: a 40-Year Perspective
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.Get Access
Saving bone by resurfacing the femoral head is not a new concept and the appeal for this type of hip replacement has remained despite the difficulties to find a bearing material suitable for this procedure.
In this article, the unique experience of a surgeon who has been performing hip resurfacing since its early development is presented, along with a comparative analysis of the performance of successive designs.
The overall 10-year Kaplan–Meier survivorship of the early designs with polyethylene bearings did not exceed 62% while that of the current Conserve®Plus metal-on-metal hybrid design implanted with second generation surgical technique is in excess of 92%. Further exceptional, in the 10-year survivorship, 99.7% has been achieved with femoral size of 46 mm and good bone quality. Cementless acetabular components provide better enduring fixation than cemented designs.
Metal-on-metal is currently and fortunately the only highly successful bearing material that can combine low wear rates and the manufacturing of a thin acetabular component to preserve bone and still accommodate the large femoral head of a hip resurfacing. The adverse local tissue reactions (ALTR) associated with metal-on-metal devices are not a bearing material issue per se but one of the device design and surgical technique. Almost all of ALTR and the rare events of systemic toxicity are due to abnormal wear patterns which can be prevented by proper acetabular component design and precise socket orientation in both the coronal and sagittal planes. Further improvements of the long-term durability with hip resurfacing can be anticipated with the use of recently developed trabecular bone-like tantalum or titanium porous coatings and with proper training of the surgeons interested in performing hip resurfacing arthroplasty.
- Amstutz, H, Beaulé, P, Dorey, F, Duff, M, Campbell, P, Gruen, T (2006) Metal-on-metal hybrid surface arthroplasty—surgical technique. J Bone and Joint Surg 88 A: pp. 234-249 CrossRef
- Amstutz, H, Grigoris, P, Dorey, F (1998) Evolution and future of surface replacement of the hip. J Orthop Sci 3: pp. 169-86 CrossRef
- Amstutz, H, Duff, M (2006) Background of metal-on-metal resurfacing. Proc Inst Mech Eng [H] 220: pp. 85-94
- Amstutz, H, Duff, M Evolution of hip resurfacing. In: Amstutz, HC eds. (2008) Hip Resurfacing: Principles, Indications, Technique and Results. Elsevier, Philadelphia, pp. 1-15
- Amstutz, H, Duff, M, Campbell, P, Dorey, F (2007) The effects of technique changes on aseptic loosening of the femoral component in hip resurfacing. Results of 600 Conserve Plus with a 3–9 year follow-up. J Arthroplasty 22: pp. 481-9 CrossRef
- Amstutz, HC Surface replacement arthroplasty. In: Amstutz, HC eds. (1991) Total Hip Arthroplasty. Churchill Livingstone, New York, pp. 295-332
- Amstutz, H. C.; Campbell, P.; Kossovsky, N.; and Clarke, I. C.: Mechanism and clinical significance of wear debris-induced osteolysis. Clin Orthop, (276): 7–18, 1992.
- Amstutz, HC, Campbell, P, Nasser, S, Kossovsky, N Modes of failure of surface replacements. In: Amstutz, HC eds. (1991) Hip Arthroplasty. Churchill Livingstone, New York, pp. 507-534
- Amstutz, H. C.; Clarke, I. C.; Christie, J.; and Graff-Radford, A.: Total hip articular replacement by internal eccentric shells: the tharies approach to total surface replacement arthroplasty. Clin Orthop, (128): 261–84., 1977.
- Amstutz, HC, Sparling, EA, Grigoris, P, Campbell, PA, Dorey, FJ (1998) Surface replacement: the hip replacement of the future. Hip International 8: pp. 187-207
- Amstutz, HC, Takamura, K, Duff, M (2011) The effect of patient selection and surgical technique on the results of Conserve® Plus hip resurfacing—3.5- to 14-year follow-up. Orthop Clin North Am I42: pp. 133-42 CrossRef
- Ball, S, Duff, M, Amstutz, H (2007) Early results of conversion of a failed femoral component in hip resurfacing arthroplasty. J Bone and Joint Surg Am 89: pp. 735-41 CrossRef
- Banerjee, M, Bouillon, B, Banerjee, C, Bäthis, H, Lefering, R, Nardini, M, Schmidt, J (2010) Sports activity after total hip resurfacing. Am J Sports Med. 38: pp. 1229-36 CrossRef
- Beaulé, P, Dorey, F, Duff, M (2003) Fate of cementless acetabular components retained during revision of a femoral component of a total hip arthroplasty. Journal of Bone and Joint Surgery 85-A: pp. 2288-93
- Beaulé, P, Duff, M, Campbell, P, Dorey, F, Park, S, Amstutz, H (2004) Metal-on-metal surface arthroplasty with a cemented femoral component: A 7–10 year follow-up study. J Arthroplasty 19: pp. 17-22
- Campbell, P.; Mirra, J.; and Amstutz, H. C.: Viability of femoral heads treated with resurfacing arthroplasty. J Arthroplasty, 15(1):120–2, 2000.
- Cooper, RA, McAllister, CM, Borden, LS, Bauer, TW (1992) Polyethylene debris-induced osteolysis and loosening in uncemented total hip arthroplasty. A cause of late failure. Journal of Arthroplasty 7: pp. 285-90 CrossRef
- Haan, R, Pattyn, C, Gill, H, Murray, D, Campbell, P, Smet, K (2008) Correlation between inclination of the acetabular component and metal ion levels in metal-on-metal hip resurfacing replacement. J Bone Joint Surg Br. 90: pp. 1291-7 CrossRef
- Eswaramoorthy, V, Biant, L, Field, R (2009) Clinical and radiological outcome of stemmed hip replacement after revision from metal-on-metal resurfacing. J Bone Joint Surg Br. 91: pp. 1454-8 CrossRef
- Garbuz, D, Tanzer, M, Greidanus, N, Masri, B, Duncan, C (2010) The John Charnley Award: Metal-on-Metal Hip Resurfacing versus Large-diameter Head Metal-on-Metal Total Hip Arthroplasty: A Randomized Clinical Trial. Clin Orthop Relat Res. 468: pp. 318-25 CrossRef
- Gruen, T, Duff, M, Wisk, L, Amstutz, H (2011) Prevalence and clinical relevance of radiographic signs of impingement in metal-on-metal hybrid hip resurfacing. J Bone and Joint Surg Am. 93: pp. 1519-26 CrossRef
- Head, WC (1982) The Wagner surface replacement arthroplasty. Orthop Clin North Am 13: pp. 789-97
- Ikeda, T, Takahashi, K, Kabata, T, Sakagoshi, D, Tomita, K, Yamada, M (2010) Polyneuropathy caused by cobalt–chromium metallosis after total hip replacement. Muscle Nerve 42: pp. 140-3 CrossRef
- Jasty, M, Bragdon, C, Jiranek, W, Chandler, H, Maloney, W, Harris, WH (1994) Etiology Of Osteolysis Around Porous-Coated Cementless Total Hip Arthroplasties. Clinical Orthopaedics and Related Research 308: pp. 111-126
- Jolley, M, Salvati, E, Brown, G (1982) Early Results and Complications of Surface Replacement of the Hip. J Bone Joint Surg [AM] 64: pp. 366-377
- Kumar, M, Shetty, M, Kiran, K, Kini, A (2012) Validation of navigation assisted cup placement in total hip arthroplasty. Int Orthop. 36: pp. 17-22 CrossRef
- Langton, D, Jameson, S, Joyce, T, Gandhi, J, Sidaginamale, R, Mereddy, P, Lord, J, Nargol, A (2011) Accelerating failure rate of the ASR total hip replacement. J Bone Joint Surg Br. 93: pp. 1011-6 CrossRef
- Langton, D, Sprowson, A, Joyce, T, Reed, M, Carluke, I, Partington, P, Nargol, A (2009) Blood metal ion concentrations after hip resurfacing arthroplasty: a comparative study of articular surface replacement and Birmingham Hip Resurfacing arthroplasties. JBJS Br 91: pp. 1287-95
- Duff, M, Amstutz, HC (2011) Sporting activity after hip resurfacing: Changes over time. Orthop Clin North Am 42: pp. 161-7 CrossRef
- Mai, MT, Schmalzried, TP, Dorey, FJ, Campbell, PA, Amstutz, HC (1996) The contribution of frictional torque to loosening at the cement–bone interface in Tharies hip replacements. J Bone Joint Surg Am 78: pp. 505-11
- Maloney, WJ, Smith, RL (1995) Periprosthetic osteolysis in total hip arthroplasty: the role of particulate wear debris. Journal of Bone and Joint Surgery. American Volume 77A: pp. 1448-1461
- Mao, X, Wong, A, Crawford, R (2011) Cobalt toxicity—an emerging clinical problem in patients with metal-on-metal hip prostheses?. Med J Aust. 194: pp. 649-51
- Murray, W. R., and Van Meter, J. W.: Surface replacement hip arthroplasty: results of the first seventy-four consecutive cases at the University of California, San Francisco. Hip: 156–66, 1982.
- Naal, F, Maffiuletti, N, Munzinger, U, Hersche, O (2007) Sports After Hip Resurfacing Arthroplasty. Am J Sports Med. 35: pp. 705-711 CrossRef
- Narvani, A, Tsiridis, E, Nwaboku, H, Bajekal, R (2006) Sporting activity following Birmingham hip resurfacing. Int J Sports Med. 27: pp. 505-7 CrossRef
- Oldenburg, M, Wegner, R, Baur, X (2009) Severe cobalt intoxication due to prosthesis wear in repeated total hip arthroplasty. J Arthroplasty 24: pp. 825.e 15-20 CrossRef
- Pazzaglia, U, Apostoli, P, Congiu, T, Catalani, S, Marchese, M, Zarattini, G (2011) Cobalt, chromium and molybdenum ions kinetics in the human body: data gained from a total hip replacement with massive third body wear of the head and neuropathy by cobalt intoxication. Arch Orthop Trauma Surg. 131: pp. 1299-308 CrossRef
- Rizzetti, M, Liberini, P, Zarattini, G, Catalani, S, Pazzaglia, U, Apostoli, P, Padovani, A (2009) Loss of sight and sound. Could it be the hip?. Lancet 373: pp. 1052 CrossRef
- Sandiford, N, Muirhead-Allwood, S, Skinner, J (2010) Revision of failed hip resurfacing to total hip arthroplasty rapidly relieves pain and improves function in the early post operative period. J Orthop Surg Res. 5: pp. 88 CrossRef
- Schmalzried, T.: Metal–metal bearing surfaces in hip arthroplasty. Orthopedics., 32(9): E-publication, 2009.
- Schmalzried, T. P.; Fowble, V. A.; and Amstutz, H. C.: The fate of pelvic osteolysis after reoperation. No recurrence with lesional treatment. Clin Orthop, (350): 128–37, 1998.
- Smith-Petersen, M (1948) Evolution of mould arthroplasty of the hip joint. J. Bone Joint Surg 30 B: pp. 59-75
- Steens, W, Foerster, G, Katzer, A (2006) Severe cobalt poisoning with loss of sight after ceramic–metal pairing in a hip—a case report. Acta Orthop. 77: pp. 830-2 CrossRef
- Su, E, Sheehan, M, Su, S (2010) Comparison of Bone Removed During Total Hip Arthroplasty With a Resurfacing or Conventional Femoral Component: A Cadaveric Study. J Arthroplasty Suppl 25: pp. 325-9 CrossRef
- Tower, S.: Arthroprosthetic Cobaltism: Neurological and Cardiac Manifestations in Two Patients with Metal-on-Metal Arthroplasty: A Case Report. J Bone Joint Surg Am., 92(7): 2847–51, 2010.
- Treacy, R, McBryde, C, Shears, E, Pynsent, P (2011) Birmingham hip resurfacing: A MINIMUM FOLLOW-UP OF TEN YEARS. J Bone Joint Surg Br. 93: pp. 27-33 CrossRef
- Vendittoli, P, Lavigne, M, Girard, J, Roy, A (2006) A randomised study comparing resection of acetabular bone at resurfacing and total hip replacement. J Bone Joint Surg Br. 88: pp. 997-1002 CrossRef
- Wera, G, Gillespie, R, Petty, C, Petersilge, W, Kraay, M, Goldberg, V (2010) Revision of hip resurfacing arthroplasty. Am J Orthop (Belle Mead NJ) 39: pp. E78-83
- Hip Resurfacing: a 40-Year Perspective
HSS Journal ®
Volume 8, Issue 3 , pp 275-282
- Cover Date
- Print ISSN
- Online ISSN
- Additional Links
- hip resurfacing
- long term
- Industry Sectors