A Mid-Term Analysis Suggests Ceramic on Ceramic Hip Arthroplasty Is Durable with Minimal Wear and Low Risk of Squeak
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.Get Access
The incidence of hip replacements in the younger patient is ever increasing. With this in mind, improving the longevity of hip arthroplasties is paramount. Alumina ceramic is a promising bearing surface due to its low wear rate and biological inertness.
This study aims to review our experience with ceramic-on-ceramic total hip arthroplasty, reporting on the need for revision as well as the cause of failure. Our secondary purpose is to review our experience with the phenomenon of squeaking analysing and its effect on clinical outcome with specific emphasis on component positioning. Also reported are the results of our retrieval analysis of explanted components documenting the wear rate and our analysis of strip wear.
A consecutive series of 301 primary cementless alumina-on-alumina total hip arthroplasties at a minimum of 10 years follow-up was reviewed. These arthroplasties all had third-generation ceramic-on-ceramic bearings performed through a posterior approach with repair of capsule and external rotators to bone. We analysed hips both clinically and radiographically. Analysis of wear in 62 ceramic bearings was performed using a Roundtest RA300 machine (Mitutoyo; Andover, UK), which has an accuracy of 0.01 μm.
Overall, the survival rate of the implants was 98% at 10 years. No ceramic fractures were encountered in this study. Seventy-four patients reported squeaking hips, and two cases were revised due to squeaking (0.6%). No failures were related to bearing wear.
We believe that ceramic-on-ceramic is a safe bearing coupling with excellent survivorship at 10 years.
- Boutin P: Total hip arthroplasty using a ceramic prosthesis. Pierre Boutin (1924-1989). Clin Rel Res 379:3-11, 2000. CrossRef
- Chevillotte C, Pibarot V, Carret JP, Bejui-Hugues J, Guyen O. Nine years follow-up of 100 ceramic-on-ceramic total hip arthroplasty. Int Orthop. 2011 Nov;35(11):1599-604. Epub 2010 Dec 21.
- D'Antonio JA, Capello WN, Naughton M. Ceramic bearings for total hip Arthroplasty have a high survivorship at 10 years. Clin Orthop Relat Res; In press.
- Dumbledon JH, Manley MT, Edidin AA. A literature review of the association between wear rate and osteolysis in total hip Arthroplasty. J Arthroplasty. 2002;17:649-61. CrossRef
- Garcia-Cimbrelo E, Martinez-Sayanes JM, Minuesa A, Minuesa L. Mittelmeier ceramic-ceramic prosthesis after 10 years. J Arthroplasty. 1996;11:773-81. CrossRef
- Hannouche D, Nich C, Bizot P, Meunier A, Nizard R, Sedel L: Fractures of ceramic bearings: history and present status. Clin Orthop Relat Res 417:19-26, 2003.
- Jarrett CA, Ranawat AS, Bruzzone M, Blum YC, Rodriguez JA, Ranawat CS: The squeaking hip: A phenomenon of ceramic-on-ceramic total hip Arthroplasty. J Bone Joint Surg Am 2009;91(6):1344-1349. CrossRef
- Kress AM, Schmidt R, Holzwarth U, Forst R, Muller LA. Excellent results with cementless total hip arthroplasty and alumina-on-alumina pairing: minimum ten-year follow-up. Int Orthop. 2011;35:195-200. CrossRef
- Kuntz M: Validation of a new high performance alumina matrix composite for use in total joint replacement. Semin Arthroplasty 2006;17:141-145. CrossRef
- Lee YK, Ha YC, Yoo JJ, Koo KH, Toon KS, Kim HJ. Alumina-on-alumina total hip Arthroplasty at minimum 10-year follow-up. J Bone Joint Surg Am. 2010;92:1715-1718. CrossRef
- Lewis PM, Al-Belooshi A, Olsen M, Schemitch EH, Waddell JP. Prospective randomized trial comparing alumina ceramic-on-ceramic with ceramic-on-conventional polyethylene bearings in total hip arthroplasty. J Arthroplasty. 2010 Apr;25(3):392-7. CrossRef
- Lusty PJ, Tai CC, Sew-Hoy RP, Walter WL, Walter WK, Zicat BA. Third-generation Alumina-on-alumina ceramic bearings in cementless total hip Arthroplasty. J. Bone Joint Surg. Am., Dec 2007; 89: 2676-2683 CrossRef
- Park YS, Park SJ, Lim SJ. Ten year results after cementless THA with sandwich-type alumina ceramic bearing. Orthopaedics.2010;33:796.
- Restrepo C, Parvizi J, Kurtz SM, Sharkey PF, Hozack WJ, Rothman RH: The noisy hip ceramic hip, is component malposition the cause? J Arthroplasty 2008;23(5):643-649. CrossRef
- Sexton SA, Yeung E, Jackson MP, Rajaratnam S, Martell JM, Walter WL, Zicat BA, Walter WK. The role of patient factors and implant position in squeaking of ceramic-on-ceramic total hip replacements.
- Sung Kwan Hwang et al. Fracture-dissociation of ceramic liner. Orthopaedics August 2008;21(8):804.
- Walter WL, Kurtz SM, Esposito C, Hozak W, Holley KG, Garino JP, Tuke MA. Retrieval analysis of squeaking alumina ceramic-on-ceramic bearings. J Bone Joint Surg Br Vol93-B, Issue 12, 1597-1601.
- Walter WL, O’Toole GC, Walter WK, Ellis A, Zicat BA: Squeaking in ceramic-on-ceramic hips: The importance of acetabular component orientation. J Arthroplasty 2007;22(4):496-503. CrossRef
- Walter WL, Waters TS, Gilles M et al: Squeaking Hips. J Bone Joint Surg Am 2008;90(suppl 4):102-111. CrossRef
- Willmann G. Ceramic femoral head retrieval data. Clin Orthop Relat Res. 2003;417:19-26.
- Yeung E, Bott PT, Chana R, Jackson MP, Holloway I, Walter WL, Zicat BA, Walter WK. Mid-term results of third-generation alumina-on-alumina ceramic bearings in cementless total hip arthroplasty: a ten-year minimum follow-up. J Bone Joint Surg Am. 2012 Jan 18;94(2):138-44 CrossRef
- Yoon TR, Rowe SM, Jung ST, Seon KJ, Maloney WJ. Osteolysis in association with a total hip arthroplasty with ceramic bearing surfaces. J Bone Joint Surg Am. 1998;80:1459-68.
- A Mid-Term Analysis Suggests Ceramic on Ceramic Hip Arthroplasty Is Durable with Minimal Wear and Low Risk of Squeak
HSS Journal ®
Volume 8, Issue 3 , pp 291-294
- Cover Date
- Print ISSN
- Online ISSN
- Additional Links
- alumina fracture
- Industry Sectors