Rationale and principle of an instrument measuring lung deposited nanoparticle surface area
Purchase on Springer.com
$39.95 / €34.95 / £29.95*
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.
The risk of nanoparticles by inhalation for human health is still being debated but some evidences of risk on specific properties of particles <100 nm diameter exist. One of the nanoparticle parameters discussed by toxicologists is their surface area concentration as a relevant property for e.g. causing inflammation. Concentrations of these small particles (~ <100 nm) are currently not measured, since the mass concentrations of these small particles are normally low despite large surface area concentrations.
Airborne particles will always be polydisperse and show a size distribution. Size is normally described by an equivalent diameter to include deviations in properties from ideal spherical particles. Here only nanoparticles below a certain size to be defined are of interest. Total concentration measures are determined by integration over the size range of interest. The ideal instrument should measure the particles according to the size weighting of the wanted quantity. Besides for the geometric surface area the wanted response function can be derived for the lung deposited surface area in the alveolar region. This can be obtained by weighting the geometric surface area as a function of particle size with the deposition efficiency for the alveolar region for e.g. a reference worker for work place exposure determination.
The investigation of the performance of an Electrical Aerosol Detector (EAD) for nearly spherical particles showed that its response function is close to the lung deposited surface areas in different regions of the human respiratory system. By changing the ion trap voltage an even better agreement has been achieved. By determining the size dependent response of the instrument as a function of ion trap voltage the operating parameters can be optimized to give the smallest error possible. Since the concept of the instrument is based on spherical particles and idealized lung deposition curves have been used, in all other cases errors will occur, which still have to be defined.
A method is now available which allows in principle the determination of the total deposited surface area in different regions of the lung in real time. It can easily be changed from one deposited region to another by varying the ion trap voltage. It has the potential to become a routine measurement technique for area measurements and personal control in e.g. work place environments.
- Donaldson K., Li X.Y., MacNee W. (1998) Ultrafine (nanometer) particle mediated lung injury. J. Aerosol Sci. 29(5–6):553–560 CrossRef
- ICRP (1994) International Commission on Radiological Protection Publication 66 Human Respiratory Tract Model for Radiological Protection. Elsevier Science Ltd, Oxford, Pergamon
- James A.C., M.R. Bailey & M-D. Dorrian, 2000. LUDEP Software, Version 2.07: Program for implementing ICRP-66 Respiratory tract model. RPB, Chilton, Didcot, OXON. OX11 ORQ UK.
- Jung H., Kittelson D. (2005) Measurement of electrical charge on diesel particles Aerosol Sci. Technol. 39(12):1129–1135 CrossRef
- Kaufman S.L., A. Medved, A. Pöcher, N. Hill, R. Caldow & F.R. Quant, 2002. An electrical aerosol detector based on the corona-jet charger. AAAR conference (poster).
- Kreyling W.G., Semmler M., Erbe F., Mayer P., Takenaka S., Schulz H., Oberdörster G., Ziesenis A. (2002) Translocation of ultrafine insoluble iridium particles from lung epithelium to extrapulmonary organs is size dependent but very low. J. Toxicol. Environ. Health Pt A 65(20):1513–1530 CrossRef
- Kruis F.E., Fissan H., Peled A. (1998) Synthesis of nanoparticles in the gas phase for electronic, optical and magnetic application – A review. J. Aerosol Sci. 29(5–6):511–535 CrossRef
- Kruis F.E. & H. Fissan, 1999. Nano-Process Technology for Synthesis and Handling of Nanoparticles. Kona Powder and Particles, No.17, pp. 130–139
- Ku B.K., Maynard A.D. (2005) Generation and investigation of airborne Ag nanoparticles with specific size and morphology by homogeneous nucleation, coagulation and sintering. J. Aerosol Sci. 36(9):1108–1124 CrossRef
- Oberdörster G., Gelein R.M., Ferin J., Weiss B. (1995) Association of particulate air pollution and acute mortality: Involvement of ultrafine particles. Inhal. Toxicol. 7:111–124
- Oberdörster G. (1996) Significance of particle parameters in the evaluation of exposure-dose–response relationships of inhaled particles. Particul. Sci. Technol. 14(2):135–151
- Rogak S.N., Flagan R.C., Nguyen H.V. (1993) The mobility and structure of aerosol agglomerates. Aerosol Sci. Technol. 18(1):25–47
- Rationale and principle of an instrument measuring lung deposited nanoparticle surface area
Journal of Nanoparticle Research
Volume 9, Issue 1 , pp 53-59
- Cover Date
- Print ISSN
- Online ISSN
- Kluwer Academic Publishers
- Additional Links
- health risk
- exposure control
- exposure measurement
- lung deposited surface area
- lung simulator
- Electrical Aerosol Detector (EAD)
- Occupational health
- Industry Sectors