Improvement of Stent Retriever Design and Efficacy of Mechanical Thrombectomy in a Flow Model
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.Get Access
In vitro experiments were performed to evaluate the efficacy of mechanical intracranial thrombectomy comparing the newly developed Aperio stent retriever and standard devices for stroke treatment.
The Aperio (A), with an increased working length of 4 cm and a special cell design for capturing and withholding clots, was compared to three benchmark devices: the Solitaire retrievable stent (B), the Merci X6 (C), and the Merci L5 retriever (D). In a vascular glass model with pulsatile flow, reminiscent of the M1 segment of the middle cerebral artery, we repeatedly induced occlusion by generating thrombi via a modified Chandler loop system. The numbers of recanalization attempts, peripheral embolizations, and recanalizations at the site of occlusion were recorded during 10 retrieval experiments with each device.
Eleven devices were able to remove the blood clots from the occluded branch. In 34 of 40 experiments, restoration of flow was obtained in 1–3 attempts. The main differences between the study devices were observed in terms of clot withholding and fragmentation during retrieval. Although there was only one fragmentation recorded for device A, disengagement of the whole clot or peripheral embolization of fragments occurred more frequently (5–7 times) with devices B, C, and D.
In a vascular model, the design of device A was best at capturing and withholding thrombi during retrieval. Further study will be necessary to see whether this holds true in clinical applications.
- Bhatia R, Hill MD, Shobha N et al (2010) Low rates of acute recanalization with intravenous recombinant tissue plasminogen activator in ischemic stroke: real-world experience and a call for action. Stroke 41:2254–2258 CrossRef
- Smith WS, Sung G, Saver J et al (2008) Mechanical thrombectomy for acute ischemic stroke: final results of the Multi MERCI trial. Stroke 39:1205–1212 CrossRef
- Kulcsar Z, Bonvin C, Pereira VM et al (2010) Penumbra system: a novel mechanical thrombectomy device for large-vessel occlusions in acute stroke. Am J Neuroradiol 31:628–633 CrossRef
- Becktepe JS, You SJ, Berkefeld J et al (2011) Clinical outcome after mechanical recanalization as mono- or adjunctive therapy in acute stroke: importance of time to recanalization. Cerebrovasc Dis 32:211–218 CrossRef
- Roth C, Papanagiotou P, Behnke S et al (2010) Stent-assisted mechanical recanalization for treatment of acute intracerebral artery occlusions. Stroke 41:2559–2567 CrossRef
- Fesl G, Patzig M, Holtmannspoetter M et al. (2011) Endovascular mechanical recanalisation after intravenous thrombolysis in acute anterior circulation stroke: the impact of a new temporary stent. Cardiovasc Interv Radiol. doi:10.1007/s00270-011-0317-5
- Castaño C, Dorado L, Guerrero C et al (2010) Mechanical thrombectomy with the Solitaire AB device in large artery occlusions of the anterior circulation: a pilot study. Stroke 41:1836–1840 CrossRef
- Machi P, Costalat V, Lobotesis K et al (2012) Solitaire FR thrombectomy system: immediate results in 56 consecutive acute ischemic stroke patients. J Neurointerv Surg 4:62–66 CrossRef
- Loh Y, Jahan R, McArthur DL (2010) Recanalization rates decrease with increasing thrombectomy attempts. AJNR Am J Neuroradiol 31:935–939 CrossRef
- Schreiber SJ, Gottschalk S, Weih M et al (2000) Assessment of blood flow velocity and diameter of the middle cerebral artery during the acetazolamide provocation test by use of transcranial doppler sonography and MR imaging. AJNR Am J Neuroradiol 21:1207–1211
- Robbie LA, Young SP, Bennett B, Booth NA (1997) Thrombi formed in a Chandler loop mimic human arterial thrombi in structure and RAI-1 content and distribution. Thromb Haemost 77:510–515
- Poole CF (1959) A study of artificial thrombi produced by a modification of Chandler’s method. Q J Exp Physiol Cogn Med Sci 44:377–384
- Liebig T, Reinartz J, Hannes R (2008) Comparative in vitro study of five mechanical embolectomy systems: effectiveness of clot removal and risk of distal embolization. Neuroradiology 50:43–52 CrossRef
- Mordasini P, Frabetti N, Gralla J (2011) In vivo evaluation of the first dedicated combined flow-restoration and mechanical thrombectomy device in a swine model of acute vessel occlusion. AJNR Am J Neuroradiol 32:294–300 CrossRef
- Improvement of Stent Retriever Design and Efficacy of Mechanical Thrombectomy in a Flow Model
CardioVascular and Interventional Radiology
Volume 36, Issue 1 , pp 192-197
- Cover Date
- Print ISSN
- Online ISSN
- Additional Links
- Acute stroke
- Interventional thrombectomy
- Stent retriever
- Industry Sectors