Comparison of the point-of-care blood gas analyzer versus the laboratory auto-analyzer for the measurement of electrolytes
Electrolyte values are measured both by arterial blood gas (ABG) analyzers and central laboratory auto-analyzers (AA), but a significant time gap exists between the availability of both these results, with the ABG giving faster results than the AA. The authors hypothesized that there is no difference between the results obtained after measurement of electrolytes by the blood gas and auto-analyzers.
After approval by the ethics committee, an observational cohort study was conducted in which 200 paired venous and arterial samples from patients admitted to the Medical Intensive Care Unit (ICU) of Apollo Hospital, Hyderabad, India, were analyzed for electrolytes on the ABG machine and the AA. Analyses were done on the ABL555 blood gas analyzer and the Dade Dimension RxL Max, both located in the central laboratory. Statistical analyses were performed using paired Student’s t test.
A total of 200 paired samples were analyzed. The mean ABG sodium value was 131.28 (SD 7.33), and the mean AA sodium value was 136.45 (SD 6.50) (p < 0.001). The mean ABG potassium value was 3.74 (SD 1.92), and the mean AA potassium value was 3.896 (SD 1.848) (p = 0.2679).
Based on the above analysis, the authors found no significant difference between the potassium values measured by the blood gas machine and the auto-analyzer. However, the difference between the measured sodium was found to be significant. We therefore conclude that critical decisions can be made by trusting the potassium values obtained from the arterial blood gas analysis.
- Price, CP (2002) Medical and economic outcomes of point of care testing. Clin Chem Lab Med 40: pp. 246-251 CrossRef
- Cox, CJ (2001) Acute care testing. Bood gases and electrolytes at the point of care. Clin Lab Med. 21: pp. 321-335
- Flegar-Mestric, Z, Perkov, S (2006) Comparability of point of care whole blood electrolyte and substrate testing using a Stat Profile Critical Care Xpress analyzer and standard laboratory methods. Clin Chem Lab Med 44: pp. 898-903 CrossRef
- Pungor, E (1992) Working mechanism of ion-selective electrodes. Pure & Appl. Chem 64: pp. 503-507 CrossRef
- The United States Clinical Laboratory Improvement Amendments
- Morimatsu, H, Rocktäschel, J, Bellomo, R, Uchino, S, Goldsmith, D, Gutteridge, G (2003) Comparison of point-of-care versus central laboratory measurement of electrolyte concentrations on calculations of the anion gap and the strong ion difference. Anesthesiology 98: pp. 1077-1084 CrossRef
- Prichard, JS, French, JS, Alvar, N (2006) Clinical evaluation of the ABL-77 for point of care analysis in the cardiovascular operating room. J Extra Corpor Technol 38: pp. 128-133
- Jacobs, E, Ancy, JJ, Smith, M (2002) Multi site performance evaluation of pH, bood gas, electrolyte, glucose and lactate determinations with the GEM Premier 3000 critical care analyzer. Point of Care 1: pp. 135-144
- Rouse C. Theoretical reasons for sodium measurement error - POCT specialist -C29-A2 ISBN 1–56238–410–4 ISSN 0273–3099
- Standardization of sodium and potassium ion-selective electrode systems to the flame photometric reference method; Approved Standard—Second Edition NCCLS.
- Lam, HS, Chan, MH, Ng, PC, Wong, W, Cheung, RC, So, AK, Fok, TF, Lam, CW (2005) Are your hands clean enough for point of care electrolyte analysis?. Pathology 37: pp. 299-304 CrossRef
- Steinfelder-Visscher, J, Weerwind, PW, Teerenstra, S, Pop, GA, Brouwer, RM (2007) Conductivity based hematocrit meaurement during cardiopulmonary bypass. J Clin Monit Comput 21: pp. 7-12 CrossRef
- Vos, G, Engel, M, Ramsay, G, Waardenburg, D (2006) Point of care blood analyzer during interhospital transport of critically ill children. Eur J Emerg Med 13: pp. 304-307 CrossRef
- Myers, GJ, Browne, J (2007) Point of care hematocrit and hemoglobin in cardiac surgery: a review. Perfusion 22: pp. 179-183 CrossRef
- Magny, E, Beaudeux, JL, Launay, JM (2003) Point care testing in blood gas and electrolyte analysis: examples of implementation and cost analysis. Ann Biol Clin (Paris) 61: pp. 344-351
- Dimeski, G, Barnett, RJ (2005) Effects of total plasma protein concentration on plasma sodium, potassium and chloride measurements by an indirect ion selective electrode measuring system. Crit Care Resusc 7: pp. 12-15
- Chow, E, Fox, N, Gama, R (2008) Effect of low serum total protein on sodium and potassium measurement by ion-selective electrodes in critically ill patients. Br J Biomed Sci 65: pp. 128-131
- Daures, MF, Combescure, C, Cristol, JP (2007) Comparitive study of siz blood gas analyzers. Ann Biol Clin (Paris) 65: pp. 505-518
- Comparison of the point-of-care blood gas analyzer versus the laboratory auto-analyzer for the measurement of electrolytes
- Open Access
- Available under Open Access This content is freely available online to anyone, anywhere at any time.
International Journal of Emergency Medicine
Volume 2, Issue 2 , pp 117-120
- Cover Date
- Print ISSN
- Online ISSN
- Additional Links