What Are the Current Clinical Issues in Wear and Tribocorrosion?
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.Get Access
Wear and corrosion in joint arthroplasty are important causes of failure. From the standpoint of current clinical importance, there are four main categories of wear and tribocorrosion: polyethylene wear, ceramic-on-ceramic (CoC) bearing wear, metal-on-metal (MoM) bearing wear, and taper tribocorrosion. Recently, problems with wear in the knee have become less prominent as have many issues with hip polyethylene (PE) bearings resulting from the success of crosslinked PE. However, MoM articulations and taper tribocorrosion have been associated with soft tissue inflammatory responses, and as a result, they have become prominent clinical concerns.
Where Are We Now?
For PE wear in the hip, several advances include improved locking mechanisms and data supporting highly crosslinked polyethylenes (HXLPE). Edge-loading in CoC articulations can contribute to stripe wear and subsequent squeaking. For MoM articulations, the relationship of wear-to-edge loading, sensitivity to component positioning, typical soft tissue response, and use of imaging is increasingly understood. Taper tribocorrosion (from femoral head–neck junctions and other modular elements) and associated soft tissue inflammatory responses appear to be serious clinical issues that are not fully understood.
Where Do We Need to Go?
In the knee, clinical concerns remain with the efficacy of HXLPE, modular connections, and metal allergies. For PE wear in the hip, concerns remain regarding how to increase crosslinking of PE while minimizing PE fractures. With CoC articulations, questions remain on how to prevent noises, chipping, and impingement and if enhanced designs can contribute to improved results. For MoM articulations, we need to improve imaging tests for soft tissue reactions, determine best practices in terms of monitoring protocols, and better define if, how, and when to act on serum metal levels. For taper tribocorrosion, we need to use modularity wisely and also understand how to improve tapers and materials in the future. For patients at risk for tribocorrosion, we need to define realistic diagnostic and monitoring protocols. We also need to enhance revision methods, and the threshold of acceptable soft tissue damage, to minimize complications associated with soft tissue damage such as hip instability.
How Do We Get There?
HXLPE and other bearing surfaces will likely continue to be refined. We need to develop tapers with more resistance to tribocorrosion through improved understanding of the manufacturing process and ongoing engineering improvements. Revision procedures for wear and tribocorrosion can be enhanced by determining when partial component retention is appropriate and how best to manage soft tissue damage. For CoC articulations, enhanced designs are required to minimize noises, chipping, and impingement. Importantly, we must continue to promote and analyze joint replacement registries to identify early failures and analyze long-term successes.
- Abdel MP, Heyse TJ, Elpers ME, Mayman DJ, Su EP, Pellicci PM, Wright TM, Padgett DE. Ceramic liner fractures presenting as squeaking after primary total hip arthroplasty. J Bone Joint Surg Am. 2014;96:27–31. CrossRef
- Amstutz HC, Le Duff MJ, Johnson AJ. Socket position determines hip resurfacing 10-year survivorship. Clin Orthop Relat Res. 2012;470:3127–3133. CrossRef
- Babovic N, Trousdale RT. Total hip arthroplasty using highly cross-linked polyethylene in patients younger than 50 years with minimum 10-year follow-up. J Arthroplasty. 2013;28:815–817. CrossRef
- Bojescul JA, Xenos JS, Callaghan JJ, Savory CG. Results of porous-coated anatomic total hip arthroplasty without cement at fifteen years: a concise follow-up of a previous report. J Bone Joint Surg Am. 2003;85:1079–1083.
- Bozic KJ, Kurtz SM, Lau E, Ong K, Chiu V, Vail TP, Rubash HE, Berry DJ. The epidemiology of revision total knee arthroplasty in the United States. Clin Orthop Relat Res. 2010;468:45–51. CrossRef
- Bozic KJ, Kurtz SM, Lau E, Ong K, Vail TP, Berry DJ. The epidemiology of revision total hip arthroplasty in the United States. J Bone Joint Surg Am. 2009;91:128–133. CrossRef
- Bragdon CR, Martell JM, Estok DM 2nd, Greene ME, Malchau H, Harris WH. A new approach for the Martell 3-D method of measuring polyethylene wear without requiring the cross-table lateral films. J Orthop Res. 2005;23:720–725. CrossRef
- Bragdon CR, Martell JM, Greene ME, Estok DM 2nd, Thanner J, Karrholm J, Harris WH, Malchau H. Comparison of femoral head penetration using RSA and the Martell method. Clin Orthop Relat Res. 2006;448:52–57. CrossRef
- Chen Z, Wang Z, Wang Q, Cui W, Liu F, Fan W. Changes in early serum metal ion levels and impact on liver, kidney, and immune markers following metal-on-metal total hip arthroplasty. J Arthroplasty. 2014;29:612–616. CrossRef
- Davies AP, Willert HG, Campbell PA, Learmonth ID, Case CP. An unusual lymphocytic perivascular infiltration in tissues around contemporary metal-on-metal joint replacements. J Bone Joint Surg Am. 2005;87:18–27. CrossRef
- Della Valle CJ, Mesko NW, Quigley L, Rosenberg AG, Jacobs JJ, Galante JO. Primary total hip arthroplasty with a porous-coated acetabular component. A concise follow-up, at a minimum of twenty years, of previous reports. J Bone Joint Surg Am. 2009;91:1130–1135.
- Hailer NP, Bengtsson M, Lundberg C, Milbrink J. High Metal ion levels after use of the ASR device correlate with development of pseudotumors and T cell activation. Clin Orthop Relat Res. 2014;472:953–961. CrossRef
- Hamilton WG, Hopper RH Jr, Engh CA Jr, Engh CA. Survivorship of polyethylene liner exchanges performed for the treatment of wear and osteolysis among porous-coated cups. J Arthroplasty. 2010;25:75–80. CrossRef
- Hannouche D, Delambre J, Zadegan F, Sedel L, Nizard R. Is there a risk in placing a ceramic head on a previously implanted trunion? Clin Orthop Relat Res. 2010;468:3322–3327. CrossRef
- Hayter CL, Gold SL, Koff MF, Perino G, Nawabi DH, Miller TT, Potter HG. MRI findings in painful metal-on-metal hip arthroplasty. AJR Am J Roentgenol. 2012;199:884–893. CrossRef
- Hayter CL, Koff MF, Potter HG. Magnetic resonance imaging of the postoperative hip. J Magn Reson Imaging. 2012;35:1013–1025. CrossRef
- Hayter CL, Koff MF, Shah P, Koch KM, Miller TT, Potter HG. MRI after arthroplasty: comparison of MAVRIC and conventional fast spin-echo techniques. AJR Am J Roentgenol. 2011;197:W405–411. CrossRef
- Hayter CL, Potter HG, Su EP. Imaging of metal-on-metal hip resurfacing. Orthop Clin North Am. 2011;42:195–205, viii.
- Hui AJ, McCalden RW, Martell JM, MacDonald SJ, Bourne RB, Rorabeck CH. Validation of two and three-dimensional radiographic techniques for measuring polyethylene wear after total hip arthroplasty. J Bone Joint Surg Am. 2003;85:505–511.
- Jacobs JJ, Urban RM, Hallab NJ, Skipor AK, Fischer A, Wimmer MA. Metal-on-metal bearing surfaces. J Am Acad Orthop Surg. 2009;17:69–76.
- Jarrett CA, Ranawat AS, Bruzzone M, Blum YC, Rodriguez JA, Ranawat CS. The squeaking hip: a phenomenon of ceramic-on-ceramic total hip arthroplasty. J Bone Joint Surg Am. 2009;91:1344–1349. CrossRef
- Kraay MJ, Moore RD, Martell JM, Rimnac CM. Reassessment of computerized wear measurement for total hip arthroplasty with correction for projectional image distortion: a brief follow-up report. J Bone Joint Surg Am. 2010;92:1858–1867. CrossRef
- Lavernia CJ, Baerga L, Barrack RL, Tozakoglou E, Cook SD, Lata L, Rossi MD. The effects of blood and fat on Morse taper disassembly forces. Am J Orthop (Belle Mead NJ). 2009;38:187–190.
- Lombardi AV Jr, Barrack RL, Berend KR, Cuckler JM, Jacobs JJ, Mont MA, Schmalzried TP. The Hip Society: algorithmic approach to diagnosis and management of metal-on-metal arthroplasty. J Bone Joint Surg Br. 2012;94:14–18. CrossRef
- Martell JM, Berkson E, Berger R, Jacobs J. Comparison of two and three-dimensional computerized polyethylene wear analysis after total hip arthroplasty. J Bone Joint Surg Am. 2003;85:1111–1117.
- McCalden RW, MacDonald SJ, Rorabeck CH, Bourne RB, Chess DG, Charron KD. Wear rate of highly cross-linked polyethylene in total hip arthroplasty. A randomized controlled trial. J Bone Joint Surg Am. 2009;91:773–782. CrossRef
- Nassif NA, Nawabi DH, Stoner K, Elpers M, Wright T, Padgett DE. Taper design affects failure of large-head metal-on-metal total hip replacements. Clin Orthop Relat Res. 2014;472:564–571. CrossRef
- Nawabi DH, Gold S, Lyman S, Fields K, Padgett DE, Potter HG. MRI predicts ALVAL and tissue damage in metal-on-metal hip arthroplasty. Clin Orthop Relat Res. 2014;472:471–481. CrossRef
- Nawabi DH, Hayter CL, Su EP, Koff MF, Perino G, Gold SL, Koch KM, Potter HG. Magnetic resonance imaging findings in symptomatic versus asymptomatic subjects following metal-on-metal hip resurfacing arthroplasty. J Bone Joint Surg Am. 2013;95:895–902. CrossRef
- Nawabi DH, Nassif NA, Do HT, Stoner K, Elpers M, Su EP, Wright T, Potter HG, Padgett DE. What causes unexplained pain in patients with metal-on metal hip devices? A retrieval, histologic, and imaging analysis. Clin Orthop Relat Res. 2014;472:543–554. CrossRef
- Niki Y, Matsumoto H, Otani T, Yoshimine F, Inokuchi W, Morisue H. Gigantic popliteal synovial cyst caused by wear particles after total knee arthroplasty. J Arthroplasty. 2003;18:1071–1075. CrossRef
- Park YS, Moon YW, Lim SJ, Yang JM, Ahn G, Choi YL. Early osteolysis following second-generation metal-on-metal hip replacement. J Bone Joint Surg Am. 2005;87:1515–1521. CrossRef
- Ranawat AS, Tsailis P, Meftah M, Koob TW, Rodriguez JA, Ranawat CS. Minimum 5-year wear analysis of first-generation highly cross-linked polyethylene in patients 65 years and younger. J Arthroplasty. 2012;27:354–357. CrossRef
- Registry NJ. 10th Annual Report. 2013. Available at: www.njrcentre.org.uk. Accessed November 22, 2013.
- Rehmer A, Bishop NE, Morlock MM. Influence of assembly procedure and material combination on the strength of the taper connection at the head–neck junction of modular hip endoprostheses. Clin Biomech (Bristol, Avon). 2012;27:77–83. CrossRef
- Shaver SM, Brown TD, Hillis SL, Callaghan JJ. Digital edge-detection measurement of polyethylene wear after total hip arthroplasty. J Bone Joint Surg Am. 1997;79:690–700.
- Stefl MD, Callaghan JJ, Liu SS, Pedersen DR, Goetz DD, Johnston RC. Primary cementless acetabular fixation at a minimum of twenty years of follow-up: a concise update of a previous report. J Bone Joint Surg Am. 2012;94:234–239. CrossRef
- Su EP. Ceramic–ceramic bearing: too unpredictable to use it regularly. HSS J. 2012;8:287–290. CrossRef
- Thomas GE, Simpson DJ, Mehmood S, Taylor A, McLardy-Smith P, Gill HS, Murray DW, Glyn-Jones S. The seven-year wear of highly cross-linked polyethylene in total hip arthroplasty: a double-blind, randomized controlled trial using radiostereometric analysis. J Bone Joint Surg Am. 2011;93:716–722.
- Van Der Straeten C, Grammatopoulos G, Gill HS, Calistri A, Campbell P, De Smet KA. The 2012 Otto Aufranc Award: The interpretation of metal ion levels in unilateral and bilateral hip resurfacing. Clin Orthop Relat Res. 2013;471:377–385. CrossRef
- Walter WL, Yeung E, Esposito C. A review of squeaking hips. J Am Acad Orthop Surg. 2010;18:319–326.
- Willert HG, Buchhorn GH, Fayyazi A, Flury R, Windler M, Koster G, Lohmann CH. Metal-on-metal bearings and hypersensitivity in patients with artificial hip joints. A clinical and histomorphological study. J Bone Joint Surg Am. 2005;87:28–36. CrossRef
- What Are the Current Clinical Issues in Wear and Tribocorrosion?
Clinical Orthopaedics and Related Research®
- Print ISSN
- Online ISSN
- Springer US
- Additional Links
- Industry Sectors