Contact Patch to Rim Distance Predicts Metal Ion Levels in Hip Resurfacing
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.Get Access
Component design, size, acetabular orientation, patient gender, and activity level have been suggested as factors leading to elevated metal ion concentrations after-on-metal hip resurfacing arthroplasty (MMHRA). The calculation of the contact patch to rim (CPR) distance integrates component size, design, and acetabular orientation and may be a good predictor of elevated metal ion levels.
We evaluated the effects and the predictive value of the CPR distance on serum cobalt (CoS) and chromium (CrS) ion levels.
We retrospectively studied 182 patients with Conserve Plus MMHRAs at a minimum of 12 months after surgery (median, 57 months; range, 12–165 months). CoS and CrS levels were analyzed using inductively-coupled plasma mass spectrometry. Multiple logistic regression was performed to determine which if any of the factors related to serum ion levels.
Patients with CPR distances of 10 mm or less had a 37-fold increased risk of having elevated CoS of 7 μg/L or higher. Similarly, these patients had an 11-fold increased risk of having elevated CrS of 7 μg/L or higher. Sex and University of California Los Angeles activity scores did not influence the postoperative CoS and CrS levels. The negative predictive value for CPR distance less than 10 mm was 99.3% for CoS greater than 7 μg/L and 98.0% for CrS greater than 7 μg/L.
Our observations suggest the CPR distance would be a useful indicator to determine which patients are at risk for elevated ion levels. Patients with CPR distances greater than 10 mm need not be monitored unless they become symptomatic.
- Allan, DG, Trammell, R, Dyrstad, B, Barnhart, B, Milbrandt, JC (2007) Serum cobalt and chromium elevations following hip resurfacing with the Cormet 2000 device. J Surg Orthop Adv. 16: pp. 12-18
- Amstutz, HC, Beaulé, PE, Dorey, FJ, Duff, MJ, Campbell, PA, Gruen, TA (2004) Metal-on-metal hybrid surface arthroplasty: two to six-year follow-up study. J Bone Joint Surg Am. 86: pp. 28-39
- Amstutz, HC, Beaulé, PE, Dorey, FJ, Duff, MJ, Campbell, PA, Gruen, TA (2006) Metal-on-metal hybrid surface arthroplasty: surgical technique. J Bone Joint Surg Am. 88: pp. 234-249
- Amstutz, HC, Duff, MJ, Campbell, PA, Wisk, LE, Takamura, KM (2011) Complications after metal-on-metal hip resurfacing arthroplasty. Orthop Clin North Am. 42: pp. 207-230 CrossRef
- Amstutz HC, Le Duff MJ, Johnson AJ. Socket position determines hip resurfacing 10-year survivorship. Clin Orthop Relat Res. 2012 Apr 18. [Epub ahead of print].
- Amstutz, HC, Thomas, BJ, Jinnah, R, Kim, W, Grogan, T, Yale, C (1984) Treatment of primary osteoarthritis of the hip: a comparison of total joint and surface replacement arthroplasty. J Bone Joint Surg Am. 66: pp. 228-241
- Anissian, HL, Stark, A, Gustafson, A, Good, V, Clarke, IC (1999) Metal-on-metal bearing in hip prosthesis generates 100-fold less wear debris than metal-on-polyethylene. Acta Orthop. Scand. 70: pp. 578-582 CrossRef
- Back, DL, Young, DA, Shimmin, AJ (2005) How do serum cobalt and chromium levels change after metal-on-metal hip resurfacing?. Clin Orthop Relat Res. 438: pp. 177-181 CrossRef
- Bergmann, G, Graichen, F, Rohlmann, A (1993) Hip joint loading during walking and running, measured in two patients. J Biomech. 26: pp. 969-990 CrossRef
- Biedermann, R, Tonin, A, Krismer, M, Rachbauer, F, Eibl, G, Stockl, B (2005) Reducing the risk of dislocation after total hip arthroplasty: the effect of orientation of the acetabular component. J Bone Joint Surg Br. 87: pp. 762-769 CrossRef
- Boardman, DR, Middleton, FR, Kavanagh, TG (2006) A benign psoas mass following metal-on-metal resurfacing of the hip. J Bone Joint Surg Br. 88: pp. 402-404 CrossRef
- Haan, R, Campbell, P, Reid, S, Skipor, AK, Smet, K (2007) Metal ion levels in a triathlete with a metal-on-metal resurfacing arthroplasty of the hip. J Bone Joint Surg Br. 89: pp. 538-541 CrossRef
- Haan, R, Campbell, PA, Su, EP, Smet, KA (2008) Revision of metal-on-metal resurfacing arthroplasty of the hip: the influence of malpositioning of the components. J Bone Joint Surg Br. 90: pp. 1158-1163 CrossRef
- Haan, R, Pattyn, C, Gill, HS, Murray, DW, Campbell, PA, Smet, K (2008) Correlation between inclination of the acetabular component and metal ion levels in metal-on-metal hip resurfacing replacement. J Bone Joint Surg Br. 90: pp. 1291-1297 CrossRef
- Smet, K, Haan, R, Calistri, A, Campbell, PA, Ebramzadeh, E, Pattyn, C, Gill, HS (2008) Metal ion measurement as a diagnostic tool to identify problems with metal-on-metal hip resurfacing. J Bone Joint Surg Am. 90: pp. 202-208 CrossRef
- Desy, NM, Bergeron, SG, Petit, A, Huk, OL, Antoniou, J (2011) Surgical variables influence metal ion levels after hip resurfacing. Clin Orthop Relat Res. 469: pp. 1635-1641 CrossRef
- Hart, AJ, Buddhdev, P, Winship, P, Faria, N, Powell, JJ, Skinner, JA (2008) Cup inclination angle of greater than 50 degrees increases whole blood concentrations of cobalt and chromium ions after metal-on-metal hip resurfacing. Hip Int. 18: pp. 212-219
- Hart, AJ, Skinner, JA, Henckel, J, Sampson, B, Gordon, F (2011) Insufficient acetabular version increases blood metal ion levels after metal-on-metal hip resurfacing. Clin Orthop Relat Res. 469: pp. 2590-2597 CrossRef
- Heisel, C, Silva, M, Skipor, AK, Jacobs, JJ, Schmalzried, TP (2005) The relationship between activity and ions in patients with metal-on-metal bearing hip prostheses. J Bone Joint Surg Am. 87: pp. 781-787 CrossRef
- Heisel, C, Streich, N, Krachler, M, Jakubowitz, E, Kretzer, JP (2008) Characterization of the running-in period in total hip resurfacing arthroplasty: an in vivo and in vitro metal ion analysis. J Bone Joint Surg Am. 90: pp. 125-133 CrossRef
- Hing, CB, Back, DL, Bailey, M, Young, DA, Dalziel, RE, Shimmin, AJ (2007) The results of primary Birmingham hip resurfacings at a mean of five years: an independent prospective review of the first 230 hips. J Bone Joint Surg Br. 89: pp. 1431-1438 CrossRef
- Kim, PR, Beaulé, PE, Dunbar, M, Lee, JK, Birkett, N, Turner, MC, Yenugadhati, N, Armstrong, V, Krewski, D (2011) Cobalt and chromium levels in blood and urine following hip resurfacing arthroplasty with the Conserve Plus implant. J Bone Joint Surg Am. 93: pp. 107-117 CrossRef
- Kwon, YM, Ostlere, SJ, McLardy-Smith, P, Athanasou, NA, Gill, HS, Murray, DW (2011) “Asymptomatic” pseudotumors after metal-on-metal hip resurfacing arthroplasty: prevalence and metal ion study. J Arthroplasty. 26: pp. 511-518 CrossRef
- Langton, DJ, Jameson, SS, Joyce, TJ, Webb, J, Nargol, AV (2008) The effect of component size and orientation on the concentrations of metal ions after resurfacing arthroplasty of the hip. J Bone Joint Surg Br. 90: pp. 1143-1151
- Langton, DJ, Sprowson, AP, Joyce, TJ, Reed, M, Carluke, I, Partington, P, Nargol, AV (2009) Blood metal ion concentrations after hip resurfacing arthroplasty: a comparative study of articular surface replacement and Birmingham Hip Resurfacing arthroplasties. J Bone Joint Surg Br. 91: pp. 1287-1295 CrossRef
- Langton, DJ, Sprowson, AP, Mahadeva, D, Bhatnagar, S, Holland, JP, Nargol, AV (2010) Cup anteversion in hip resurfacing: validation of EBRA and the presentation of a simple clinical grading system. J Arthroplasty. 25: pp. 607-613 CrossRef
- MacDonald, SJ, McCalden, RW, Chess, DG, Bourne, RB, Rorabeck, CH, Cleland, D, Leung, F (2003) Metal-on-metal versus polyethylene in hip arthroplasty: a randomized clinical trial. Clin Orthop Relat Res. 406: pp. 282-296 CrossRef
- McMinn, D, Treacy, R, Lin, K, Pynsent, P (1996) Metal on metal surface replacement of the hip: experience of the McMinn prosthesis. Clin Orthop Relat Res. 329: pp. S89-S98 CrossRef
- MHRA. Medical Device Alert: All metal-on-metal (MoM) hip replacements (MDA/2012/008). http://www.mhra.gov.uk/home/groups/dts-bs/documents/medicaldevicealert/con143787.pdf. Accessed September 18, 2012.
- Morlock, MM, Bishop, N, Zustin, J, Hahn, M, Ruther, W, Amling, M (2008) Modes of implant failure after hip resurfacing: morphological and wear analysis of 267 retrieval specimens. J Bone Joint Surg Am. 90: pp. 89-95 CrossRef
- Moroni, A, Savarino, L, Cadossi, M, Baldini, N, Giannini, S (2008) Does ion release differ between hip resurfacing and metal-on-metal THA?. Clin Orthop Relat Res. 466: pp. 700-707 CrossRef
- Moroni, A, Savarino, L, Hoque, M, Cadossi, M, Baldini, N (2011) Do ion levels in hip resurfacing differ from metal-on-metal THA at midterm?. Clin Orthop Relat Res. 469: pp. 180-187 CrossRef
- Pandit, H, Glyn-Jones, S, McLardy-Smith, P, Gundle, R, Whitwell, D, Gibbons, CL, Ostlere, S, Athanasou, N, Gill, HS, Murray, DW (2008) Pseudotumours associated with metal-on-metal hip resurfacings. J Bone Joint Surg Br. 90: pp. 847-851 CrossRef
- Pattyn, CA, Lauwagie, SN, Verdonk, RC (2011) Whole blood metal ion concentrations in correlation with activity level in three different metal-on-metal bearings. J Arthroplasty. 26: pp. 58-64 CrossRef
- Savarino, L, Granchi, D, Ciapetti, G, Cenni, E, Nardi Pantoli, A, Rotini, R, Veronesi, CA, Baldini, N, Giunti, A (2002) Ion release in patients with metal-on-metal hip bearings in total joint replacement: a comparison with metal-on-polyethylene bearings. J Biomed Mater Res. 63: pp. 467-474 CrossRef
- Schmalzried TP. Metal-metal bearing surfaces in hip arthroplasty. Orthopedics. 2009;32: pii: orthosupersite.com/view.asp?rID = 42831. doi: 10.3928/01477447-20090728-06.
- Steens, W, Foerster, G, Katzer, A (2006) Severe cobalt poisoning with loss of sight after ceramic-metal pairing in a hip: a case report. Acta Orthop. 77: pp. 830-832 CrossRef
- Tiberi, JV, Pulos, N, Kertzner, M, Schmalzried, TP (2012) A more reliable method to assess acetabular component position. Clin Orthop Relat Res. 470: pp. 471-476 CrossRef
- Tower, SS (2010) Arthroprosthetic cobaltism: neurological and cardiac manifestations in two patients with metal-on-metal arthroplasty. A case report. J Bone Joint Surg Am. 92: pp. 2847-2851 CrossRef
- Underwood, RJ, Zografos, A, Sayles, RS, Hart, A, Cann, P (2012) Edge loading in metal-on-metal hips: low clearance is a new risk factor. Proc Inst Mech Eng H. 226: pp. 217-226
- Vendittoli, PA, Mottard, S, Roy, AG, Dupont, C, Lavigne, M (2007) Chromium and cobalt ion release following the Durom high carbon content, forged metal-on-metal surface replacement of the hip. J Bone Joint Surg Br. 89: pp. 441-448 CrossRef
- Witzleb, WC, Ziegler, J, Krummenauer, F, Neumeister, V, Guenther, KP (2006) Exposure to chromium, cobalt and molybdenum from metal-on-metal total hip replacement and hip resurfacing arthroplasty. Acta Orthop. 77: pp. 697-705 CrossRef
- Contact Patch to Rim Distance Predicts Metal Ion Levels in Hip Resurfacing
Clinical Orthopaedics and Related Research®
Volume 471, Issue 5 , pp 1615-1621
- Cover Date
- Print ISSN
- Online ISSN
- Additional Links
- Industry Sectors
- Author Affiliations
- 1. Joint Replacement Institute at St. Vincent Medical Center, 2200 West Third Street, Suite 400, Los Angeles, CA, 90057, USA
- 2. J. Vernon Luck Sr. MD, Orthopaedic Research Center, Orthopaedic Hospital at UCLA, Los Angeles, CA, USA