Fixation Failures of Dual Mobility Cups: A Mid-term Study of 2601 Hip Replacements
- First Online:
- Cite this article as:
- Massin, P., Orain, V., Philippot, R. et al. Clin Orthop Relat Res (2012) 470: 1932. doi:10.1007/s11999-011-2213-3
- 477 Views
The use of dual-mobility cups has increased because of a low rate of dislocations combined with a 96% 15-year survival rate. However, late cup migrations have been attributed to their fixation (tripod - exact fit with two pegs and one extraacetabular screw) and the absence of porous coating. In a second-generation device, the designs were modified to achieve press-fit fixation and a layer of titanium beads was sintered on stainless steel cups.
We therefore (1) determined the midterm survival of press-fit, grit-blasted, second-generation cups with or without additional screws, compared with original tripod and (2) compared survival of grit-blasted dual-mobility cups with bimetallic porous-coated cups.
From a multiinstitutional trial, we reviewed 2408 patients with osteoarthritis implanted with 2601 prostheses of seven designs of a second-generation dual-mobility cup. The criteria for failure were migration, widening radiolucencies in any zone of the interface, or revision for cup loosening. The minimum followup was 5 years (mean, 7.7 years; range, 5–11 years).
The 8-year survival rate of press-fit, grit-blasted cups was lower than that for press-fit, grit-blasted cups fixed with screws (91% versus 100%) and for tripod fixation (98%). The 8-year survival rate of press-fit, grit-blasted cups was less than that for press-fit, porous-coated cups made of the same alloy (91% versus 95%).
The data suggested primary fixation of grit-blasted dual-mobility cups should be secured with screws. Porous coating sintered on the convex side improved midterm survivorship. No deleterious effect of metallosis resulted from sintered titanium beads on stainless steel. Long-term followup is required to confirm these findings.
Level of Evidence
Level IV, therapeutic study. See Guidelines for Authors for a complete description of levels of evidence.
Farizon et al.  introduced the concept of mobile bearings for THA in the late 1970s to reduce the risk of dislocation. The concept was reported by others [14, 17, 26]. Contrary to conventional fixed-bearing cups, the mobile polyethylene insert, designed to be head retaining, is left free to rotate in its metallic shell. In extreme movements provoking femoral neck impingement, inserts behave like a large-diameter head and are more difficult to dislocate than a conventional femoral head.
Second-generation cups became available in the late 1990s and were designed to correct the concerns of first-generation shells by replacing their alumina coating with hydroxyapatite [18, 30]. The design and outer surfaces were modified to achieve press-fit fixation followed by bone ingrowth; the inner surface had to permit low friction with the convex side of the mobile polyethylene insert, which required proper surface finish and tribologic properties. Some shells were designed for impaction alone (Fig. 1), whereas others incorporated extra fixation devices, such as flanges (Fig. 1C), allowing insertion of additional extraarticular screws to secure fixation if needed. From a metallurgic point of view, titanium was not recommended for femoral heads because of its poor tribologic properties affecting polyethylene bearings . Therefore, cobalt-chromium and stainless steel were judged to be more appropriate for mobile-bearing shells, although the manufacturing techniques available in France did not allow sintering their surfaces with beads made of the same alloy. Therefore, although most shells had grit-blasted surfaces, manufacturers coated some stainless steel cups with titanium beads. These variations allowed seven types of shells: (1) press-fit grit-blasted stainless steel, (2) press-fit grit-blasted cobalt-chromium, (3) press-fit porous-coated stainless steel, (4) press-fit grit-blasted stainless steel with screws, (5) press-fit porous-coated stainless steel with screws, (6) tripod grit-blasted, and (7) tripod porous-coated.
Our purposes were to (1) determine the mid-term survival of press-fit, grit-blasted second-generation cups with and without extraarticular screws compared with original tripod, and (2) compare survival of second-generation dual-mobility cups in terms of fixation for grit-blasted and bimetallic porous-coated cups.
Patients and Methods
We retrospectively reviewed all 3210 patients who underwent 3474 THAs between 1998 and 2003 in 16 French centers. Of these, 2408 patients underwent 2601 THAs for primary osteoarthritis. To provide a more homogeneous group, we reviewed only patients with primary osteoarthritis. With that particular etiology, indications for THA using a dual-mobility cup were: (1) advanced osteoarthritis involving complete joint narrowing and disabling pain, but conservation of some walking abilities, and (2) patients older than 55 years. The contraindications were: (1) mild osteoarthritis with incomplete joint narrowing, (2) patients younger than 55 years, and (3) patients unable to walk. Mean age was 72 ± 9 years (range, 55–92 years). There were 1429 women and 979 men. Although we asked physicians to recall patients for this study, 650 patients (696 hips or 27%) had less than 5 years followup: 416 patients (17%) (436 hips) died and 234 patients (10%) (264 hips) were lost to followup or had incomplete data. This left 1758 patients (1905 hips) with a minimum followup of 5 years (mean, 7.7 years; range, 5–11 years).
Demographics of different groups according to cup type in patients with primary osteoarthritis
Number of hips
Mean age (years)
Sex ratio female/male
(p < 0.001)
Charnley Class A/B/C
(p < 0.001)
Press-fit stainless steel
72 ± 10
26 ± 5
75 ± 8
26 ± 4
71 ± 7
27 ± 5
Press-fit stainless steel with additional screw fixation
72 ± 10
26 ± 4
Press-fit porous-coated with additional screws
74 ± 8
26 ± 5
72 ± 8
26 ± 4
Tripod porous- coated
71 ± 10
26 ± 5
Nine hundred ninety-seven cups (929 patients) designed for press-fit fixation alone had a peripheral rim for equatorial press fit. An 80-μm hydroxyapatite layer (Collegia™, Wright Medical EMEA, Amstelveen, the Netherlands) covered 402 press-fit, grit-blasted stainless steel cups (n = 91) or a bilayer of hydroxyapatite and alumina (Novae Sunfit™, Serf, Decines, France) (n = 311). Four hundred four stainless steel cups were porous coated with titanium beads, giving 150 to 400 μm diameter pores (Polarcup™, Smith & Nephew Orthopaedics AG, Rotkreuz, Switzerland [n = 301]; Saturne™, Amplitude, Valence, France [n = 103]). One hundred ninety-one cups were made of cobalt-chromium and covered with a 100-μm hydroxyapatite layer (Evora™, Science et Médecine, Montrouge, France). They were grit-blasted and supplied with rotational stabilizers (fins or small spikes).
A second group of 726 stainless steel cups (680 patients) was supplied with flanges, allowing the insertion of additional extraarticular screws. Five hundred ninety cups were grit blasted (Gyros, DePuy, Warsaw, IN, USA), while 136 were porous coated, with a plasma-sprayed layer of titanium beads (Fig. 1C [Polarcup™, Smith & Nephew Orthopaedics AG, Memphis, TN, USA]).
Finally, a third group of 878 cups (799 patients) were designed for tripod fixation according to the original model of Bousquet, as described by Farizon et al. , in which the majority had a grit-blasted surface covered with an hydroxyapatite-alumina bilayer (797 cups [Novae, Serf, Decines, France]) (Fig. 1A), and the rest were covered with a porous surface (81 cups [EOL™, Ceramconcept, Paris, France]).
Prostheses were implanted through a posterolateral approach. The acetabulum was prepared using reamers of growing size and press fit was achieved by implanting the cup with a 2-mm greater diameter than that of the last reamer. At the end of the operation, repair of the external rotator muscles was performed in 10% of the cases. Additional screw fixation then was performed in implants with flanges.
Patients were allowed immediate weightbearing as tolerated using crutches. No other physiotherapy other than walking assistance was provided during the hospital stay. We typically discharged patients between the sixth and eighth postoperative days; patients either were sent home or to a convalescent home until they recovered independence. They were encouraged to avoid high flexion combined with adduction of the involved hip for 6 weeks postoperatively, after which we recommended they stop using their crutches and resume their usual activities.
Patients were clinically examined for walking distance, pain, and motilities 6 weeks, 6 months, and yearly after surgery. Early and late complications were recorded and classified according to Dindo et al. . At each visit an AP radiograph of the pelvis was taken with the lower limbs internally rotated and including the entire implant. Magnification was calculated by the ratio between the measured and the known cup diameter. We compared these radiographs with the immediate postoperative radiographs taken using the same technique.
There were missing data for the 650 patients whose followup was less than 5 years. However, we included these patients in the first intervals of the survivor analysis (0 to 5 years) because four of these patients had early cup mobilization, which was included in the radiographic failure end point.
Six of us (VO, RP, and four observers who also were the treating physicians) independently evaluated all radiographs for cup migration and widening radiolucencies in any zone of the fixation interface. We measured cup migration on comparable AP pelvic views according to criteria established by Massin et al. , based on a minimum 5-mm variation in perpendicular vertical distance between the obturator line and the teardrop line for 10% magnification. Compared with the gold standard, radiostereometry, such manual measurements on selected radiographs provided an accuracy of 4.4 mm, with an interobserver error of 3.2 mm . We evaluated radiolucencies in the three zones of DeLee and Charnley  with the following criteria for lucencies: (1) radiolucent lines greater than 2 mm in width, (2) developing at least in acetabular Zone 3 of DeLee and Charnley, and (3) they were not present on immediate postoperative radiographs. We defined mechanical radiographic failure as cup center migration in either direction greater than 3 mm and/or variations of cup inclination greater than 10°, and/or by widening and extensive radiolucencies after the first postoperative year. We examined AP and lateral hip radiographs for osteolysis.
We performed Kaplan-Meier survivorship analysis  based on successive 1-year intervals (life table), until the latest interval including a minimum of 10 patients at risk, and using radiographic failure as a primary end point. Secondary end points were revision for any cause and revision for failure of fixation. We calculated the 95% CI according to Peto et al. . We compared 8-year survival of grit-blasted cups with different modes of primary fixation, but the same outer surface: tripod, press-fit alone, and press-fit with additional extraarticular screws. We compared survival of cups with the same primary fixation type, but different surfaces (grit-blasted versus porous-coating) to address whether outer cup surface influenced biologic fixation. Thus, we compared survival of press-fit, grit-blasted stainless steel cups with press-fit, porous-coated cups made of the same alloy, and of tripod grit-blasted cups with tripod porous-coated cups. We compared survivor curves using the log-rank test . We used version 12 of the SPSS® software (SPSS, IBM® Inc, Chicago, IL, USA) for statistical analysis.
Classification of complications according to Dindo et al.  (of a total of 2601 hips)
Regressive sciatic palsies
Leg length discrepancy 1 cm or greater
Grade II (nonrevised)
Symptomatic deep venous thrombosis
Grade III (revised)
Early dislocations (reduction under GA)
Grade IV (life threatening)
Symptomatic pulmonary embolism
Ischemic stroke/brain hemorrhage
Grade V (death)
Cumulative survival rates of different dual-mobility cup groups (%)
Press-fit, grit-blasted cobalt-chromium (n = 191)
Press-fit, grit-blasted stainless steel (n = 402) (± 95% CI)
Press-fit grit-blasted stainless steel with screws (n = 590)
Press-fit porous-coated stainless steel (n = 404) (± 95% CI)
Press-fit porous-coated stainless steel with screws (n = 136) (± 95% CI)
Tripod grit-blasted (n = 797) (± 95% CI)
Tripod porous-coated (n = 81) (± 95% CI)
97% ± 3%
99% ± 2%
99% ± 1%
96% ± 4%
98% ± 3%
99% ± 1%
98% ± 4%
91% ± 6%
98% ± 3%
91% ± 16%
98% ± 2%
90% ± 14%
84% ± 13%
98% ± 2%
98% ± 5%
p value (comparison grit-blasted/tripod)
p value (comparison grit-blasted/porous-coated
The outer surface effect (grit-blasted versus porous-coated) affected the midterm fixation of press-fit dual-mobility cups. The 8-year cumulative survival of press-fit, grit-blasted cups (with or without screws) was lower (p = 0.002) than that of porous-coated cups (with or without screws): 91% versus 95%, respectively. In contrast, tripod grit-blasted cups were fixed more durably (p = 0.04) than porous-coated cups: 98% versus 90% 8-year survival, respectively.
Second-generation dual-mobility cups were expected to correct the defaults of the first-generation tripod cups described by Farizon et al. . The delamination of the alumina layer coated on a grit-blasted surface jeopardized the fixation of the initial design. The first modification consisted of substitution of the alumina with hydroxyapatite that initially was coated on grit-blasted surfaces. At the same time, tripod fixation was converted to a press-fit fixation and eventually secured with additional extraacetabular screws. Later, titanium plasma-sprayed beads were sintered on the outer surface of stainless steel cups. To assess the modified devices we (1) determined the mid-term survival of press-fit, grit-blasted second-generation cups with or without additional screws compared with original tripod cups, and (2) compared survival of grit-blasted dual-mobility cups with bimetallic porous-coated cups.
Our study has some limitations. First, because it was a large multicenter study, it involved numerous surgeons using different techniques. Although this introduced heterogeneity, it likely enhanced the generalizability of the data. Second, we lost a substantial number of patients to followup who were included in the early intervals of the survival analysis, but which decreased the number of patients at risk after 5 years postoperatively. Nonetheless, owing to the large number of cases, we observed differences between the implant types, some of them related to the type of primary fixation, others to the type of outer surface; however, we cannot say whether the differences we observed would apply to the entire group. Third, four late failures impacted the long-term survival of tripod grit-blasted cups, which we did not consider in the survival comparisons that stopped before 9 years because the number of patients at risk in the other groups of implants was too low. Such late failures indicated either suboptimal fibrous biologic fixation, late osteolysis, or both.
Thus, we believe additional extraarticular screws are required to secure the fixation of grit-blasted stainless steel cups to improve their mid-term survival. This kind of late fixation failure is not specific to mobile-bearing cups, but was reported with fixed-bearing cementless cups without porous coating [15, 16, 24, 28]. It is explained as mechanical failure of the fixation interface independent of particle-induced bone resorption and may be attributed to the lack of microinterlocking between bone and metal in the absence of porous coating. Dual-mobility cobalt-chromium cups had the best survival rate among press-fit cups. They had reliefs or macrostructures, such as grooves, pegs, or wings implanted on their dome, whereas the stainless steel cups had only some rough texture located at their periphery . Thus, it appears that better survival with cobalt-chromium cups was attributable to their design rather than their metallurgy.
Comparisons of our mechanical fixation failure rates with those of fixed-bearing cementless cups
Grist-blasted cups % of mechanical fixation failures
Porous-coated cups % of mechanical fixation failures
Mean followup (years) or intervals of followup
Reikeras and Gunderson 
Lai et al. 
Miyakawa et al. 
Kim et al. 
Malchau et al. 
Bohm and Bosche 
Manley et al. 
D’Lima et al. 
Bojescul et al. 
Engh et al. 
Garavaglia et al. 
Vicente et al. 
Fixation of second-generation dual-mobility cups still needs improvement. Press-fit, grit-blasted cups without additional screw fixation did not provide satisfactory midterm stability and should not be used. Macrostructures fixed on their dome may be a solution to optimize rotational stability. Titanium porous coating applied on the surface of press-fit cups appeared to be promising, but did not improve the fixation of tripod cups. Thus, it seems that the best design for receiving the porous layer would be press-fit hemispherical cups. Followup beyond 10 years is required before confirming the tolerance of bialloy titanium, stainless steel cups.
This work is a multicenter retrospective study conducted in 16 French centers: Bichat Claude Bernard Hospital, Paris Diderot University, France (V. Orain, P. Massin), North Hospital, Saint Etienne University, France (R. Philippot, F. Farizon), Lyon Sud Hospital, Lyon University, France (M. Fessy), Edouard Herriot Hospital, Lyon University, France (O. Guyen), Hautepierre Hospital, Strasbourg University, France (P. Adams, P. Corcos, M. Ehlinger), North Hospital, Amiens University, France (P. Mertl, F. Leiber), Rangueil Hospital, Toulouse University, France, Orthopaedic Department (J.M. Laffosse, J. Puget), Nancy University, France (O. Roche), Saint Roch Hospital, Cavaillon, France (P. Tracol), La Meynard Hospital, Fort de France University, France (J.L. Rouvillain), Côte de Nacre Hospital, Caen University, France (G. Burdin), Saint Martin Hospital, Caen (S. Leclercq), Salengro Hospital, Lille University, France (A. Combes, J. Girard, H. Migaud), Saint Georges Hospital, Nice, France (L. Descamps, G. Dehri), South Hospital, Rennes University, France (N. Barba, D. Huten), Saint Vincent de Paul Hospital, Bourgoin Jallieu, France (D. Noyer). We thank Drs P. Adams, N. Barba, G. Burdin, A. Combes, P. Corcos, L. Descamps, G. Dehri, M. Ehlinger, J. Girard, O. Guyen, S. Leclercq, D. Noyer, O. Roche, and P. Tracol for reviewing the patients and reading the radiographs. We also thank Ovid Da Silva for editing this manuscript, and the department of biostatistics (Lille University, Lille, France, Pr Duhamel) for the statistical analysis.