Techniques and parameters for estimating radiation exposure and dose in cardiac computed tomography
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.Get Access
With increasing clinical use of cardiac CT imaging it is important that all health care providers referring for or administering such examinations are familiar with the concepts and values of radiation dosimetry in CT as well as with the basic principles of radiation protection. There are important technical differences pertinent to radiation dose between the CT scanner types that are currently being used for imaging of the heart and coronary arteries. As a result of these differences, the radiation dose typically is higher when a cardiac examination is performed with multidetector-row CT (MDCT) than when it is performed with electron beam CT. Several techniques have been described to reduce radiation dose of MDCT imaging by varying the X-ray tube current during a CT examination. The volume computed tomographic dose index (CTDIvol), the dose length product (DLP), and the effective dose (E) are the most useful parameters to describe and compare radiation doses received from cardiac CT examinations. When comparing radiation doses between scanning protocols and scanner types, the degree of image noise must be considered. Diagnostic, rather than aesthetic, quality of images should be the most important factor guiding the development of scanning protocols for cardiac CT imaging. Cardiac CT examinations should be ordered only by qualified health care providers, and the ordering clinicians should be aware of their responsibility of weighing risks of the radiation exposure against the expected benefits.
- Gerber, TC, Kuzo, RS, Karstaedt, N, Lane, GE, Morin, RL, Sheedy, PF, Safford, RE, Blackshear, JL, Pietan, JH (2002) Current results and new developments of coronary angiography with use of contrast-enhanced computed tomography of the heart.. Mayo Clin Proc 77: pp. 55-71
- Ropers, D, Ulzheimer, S, Wenkel, E, Baum, U, Giesler, T, Derlien, H, Moshage, W, Bautz, WA, Daniel, WG, Kalender, WA, Achenbach, S (2001) Investigation of aortocoronary artery bypass grafts by multislice spiral computed tomography with electrocardiographic-gated image reconstruction.. Am J Cardiol 88: pp. 792-795 CrossRef
- Deibler, AR, Kuzo, RS, Vöhringer, M, Page, EE, Safford, RE, Patton, JN, Lane, GE, Morin, RL, Gerber, TC (2004) Imaging of congenital coronary anomalies with multislice spiral computed tomography.. Mayo Clin Proc 79: pp. 1017-1023 CrossRef
- Morin, RL, Gerber, TC, McCollough, CH (2003) Radiation dose in computed tomography of the heart.. Circulation 107: pp. 917-922
- Berrington de Gonzalez, A, Darby, S (2004) Risk of cancer from diagnostic X-rays: estimates for the UK and 14 other countries.. Lancet 363: pp. 345-351
- Jakobs, TF, Becker, CR, Ohnesorge, B, Flohr, T, Suess, C, Schoepf, UJ, Reiser, MF (2002) Multislice helical CT of the heart with retrospective ECG gating: reduction of radiation exposure by ECG-controlled tube current modulation.. Eur Radiol 12: pp. 1081-1086 CrossRef
- Hunold, P, Vogt, FM, Schmermund, A, Debatin, JF, Kerkhoff, G, Budde, T, Erbel, R, Ewen, K, Barkhausen, J (2003) Radiation exposure during cardiac CT: effective doses at multi-detector row CT and electron-beam CT.. Radiology 226: pp. 145-52
- Trabold, T, Buchgeister, M, Kuttner, A, Heuschmid, M, Kopp, AF, Schroder, S, Claussen, CD (2003) Estimation of radiation exposure in 16-detector row computed tomography of the heart with retrospective ECG-gating.. ROFO-Fortschritte auf dem Gebiet der Röntgenstrahlen und der Bildgebenden V 175: pp. 1051-1055
- McCollough, CH, Zink, FE (1999) Performance evaluation of a multi-slice CT system.. Med Phys 26: pp. 2223-2230 CrossRef
- Boyd, DP, Lipton, MJ (1983) Cardiac computed tomography.. Proc IEEE 71: pp. 298-307 CrossRef
- McCollough, CH, Zink, FE, Morin, RL (1994) Radiation dosimetry for electron beam CT.. Radiology 192: pp. 637-643
- GE Healthcare. Computed tomography: E-speed. Available at: http://www.gemedicalsystemseurope.com/euen/rad/ct/products/espeed.html. Accessed June 28, 2004.
- Siemens Medical Systems. Shifting Paradigms in CT. Available at: http://www.medical.siemens.com/siemens/en_US/rg_marcom_FBAs/files/brochures/magazi_medical_ solutions/science/Science_Artikel_Speed4D.pdf. Accessed June 28, 2004.
- Shope, TB, Gagne, RM, Johnson, GC (1981) A method for describing the doses delivered by transmission X-ray computed tomography.. Med Phys 8: pp. 488-495 CrossRef
- Cameron, JR, Zimmerman, D, Kenney, G, Buch, R, Bland, R, Grant, R (1964) Thermoluminescent radiation dosimetry utilizing LiF.. Health Phys 10: pp. 25-29 CrossRef
- Suzuki, A, Suzuki, MN (1978) Use of a pencil-shaped ionization chamber for measurement of exposure resulting from a computed tomography scan.. Med Phys 5: pp. 536-539
- Pavlicek, W, Horton, J, Turco, R (1979) Evaluation of the MDH Industries, Inc. pencil chamber for direct beam CT measurements.. Health Phys 37: pp. 773-774
- Jucius, RA, Kambic, GX (1980) Measurements of computed tomography X-ray fields utilizing the partial volume effect.. Med Phys 7: pp. 379-382
- European guidelines on quality criteria for computed tomography. Available at: http://www.drs.dk/guidelines/ct/quality/mainindex.htm. Accessed June 28, 2004.
- Medical Electrical Equipment. Part 2-44: Particular Requirements for the Safety of X-ray Equipment for Computed Tomography.. International Electrotechnical Commission (IEC) Central Office, Geneva, Switzerland
- 1990 recommendations of the International Commission on Radiological Protection. Ann ICRP 1991; 21: 1–201.
- McCollough, CH, Schueler, BA (2000) Calculation of effective dose.. Med Phys 27: pp. 828-837
- Morin, RL (1988) Monte Carlo Simulation in the Radiological Sciences.. CRC Press, Boca Raton, FL
- Jones, DG, Shrimpton, P. (1991) Survey of CT Practice in the U.K. Part 3: Normalised Organ Doses Calculated Using Monte Carlo Techniques. NRPB 250.. National Radiological Protection Board, Oxon
- Zankl, M, Panzer, W, Drexler, G (1991) The Calculation of Dose from External Photon Exposures Using Reference Human Phantoms and Monte Carlo Methods. Part IV: Organ Dose From Computed Tomographic Examinations. GSF-Bericht 30/91.. GSF–Forschungszentrum für Umwelt und Gesundheit, Institut für Strahlenschutz, Neuherberg
- McCollough, CH (2003) Patient dose in cardiac computed tomography.. Herz 28: pp. 1-6
- Kalender, WA, Wolf, H, Suess, C, Gies, M, Greess, H, Bautz, WA (1999) Dose reduction in CT by on-line tube current control: principles and validation on phantoms and cadavers.. Eur Radiol 9: pp. 323-328 CrossRef
- Gies, M, Kalender, WA, Wolf, H, Suess, C (1999) Dose reduction in CT by anatomically adapted tube current modulation. I. Simulation studies.. Med Phys 26: pp. 2235-2247 CrossRef
- Kalender, WA, Wolf, H, Suess, C (1999) Dose reduction in CT by anatomically adapted tube current modulation. II. Phantom measurements.. Med Phys 26: pp. 2248-2253 CrossRef
- Gerber TC, Stratman BP, Kuzo R, Morin RL. Effect of acquisition technique on radiation dose and image quality in computed tomographic coronary angiography with submillimeter collimation. Invest Radiol (in press).
- Gerber, TC, Kuzo, R, Safford, RE (2005) Posterior Mitral Valve Leaflet Prolapse Diagnosed With Multislice Computed Tomography.. Heart 91: pp. 130
- International Commission on Radiological Protection. Managing patient dose in computed tomography. Available at: http://www.icrp.org/docs/ICRP_87_CT_s.pps. Accessed June 28, 2004.
- Techniques and parameters for estimating radiation exposure and dose in cardiac computed tomography
The International Journal of Cardiovascular Imaging
Volume 21, Issue 1 , pp 165-176
- Cover Date
- Print ISSN
- Online ISSN
- Kluwer Academic Publishers
- Additional Links
- computed tomography
- coronary artery disease
- radiation dose
- Industry Sectors