Automatic detection of spermatozoa for laser capture microdissection
- First Online:
- Cite this article as:
- Vandewoestyne, M., Van Hoofstat, D., Van Nieuwerburgh, F. et al. Int J Legal Med (2009) 123: 169. doi:10.1007/s00414-008-0271-1
- 886 Downloads
In sexual assault crimes, differential extraction of spermatozoa from vaginal swab smears is often ineffective, especially when only a few spermatozoa are present in an overwhelming amount of epithelial cells. Laser capture microdissection (LCM) enables the precise separation of spermatozoa and epithelial cells. However, standard sperm-staining techniques are non-specific and rely on sperm morphology for identification. Moreover, manual screening of the microscope slides is time-consuming and labor-intensive. Here, we describe an automated screening method to detect spermatozoa stained with Sperm HY-LITER™. Different ratios of spermatozoa and epithelial cells were used to assess the automatic detection method. In addition, real postcoital samples were also screened. Detected spermatozoa were isolated using LCM and DNA analysis was performed. Robust DNA profiles without allelic dropout could be obtained from as little as 30 spermatozoa recovered from postcoital samples, showing that the staining had no significant influence on DNA recovery.
KeywordsForensic science Sexual assault Laser capture microdissection Spermatozoa Automatic cell recognition
In cases of sexual assault, DNA typing, using biological material such as vaginal smears is a common method for suspect identification. The spermatozoa are traditionally separated from the vaginal epithelial cells by differential extraction. The epithelial cells are preferentially lysed by incubation in an SDS/proteinase K mixture, while spermatozoa are resistant to this treatment and can be separated by centrifugation . The remaining supernatant containing the victim’s DNA is removed and the spermatozoa can be lysed in a buffer containing dithiothreitol (DTT). However, a certain amount of vaginal cells remain undigested during the initial steps so that if there are only a few spermatozoa on the microscope slide, the DNA of the perpetrator is contaminated by the victim’s DNA, resulting in a mixed DNA profile  hereby making this method ineffective. In some cases the profile of the perpetrator cannot be detected due to the large amount of victim’s DNA.
Another option is the use of Y-chromosome short tandem repeats (Y STR) to detect the male component in mixed stains when the DNA of the male contributor is present in a small amount . Since the discrimination power of Y STR analysis is much lower than autosomal STR analysis, it is only of interest in cases where no autosomal DNA profile of the semen contributor could be detected.
The use of laser capture microdissection (LCM) has been evaluated for the isolation of spermatozoa from microscope slides containing both spermatozoa and vaginal cells [4, 5, 6]. Elliot et al. showed that LCM outperforms differential extraction for the recovery of DNA from sperm cells, especially in cases where only a few sperm cells are present . LCM was also used for the isolation of male cells in a male/female cell mixture [8, 9].
The search for spermatozoa on a microscope slide by visual inspection can often be very time-consuming and labor-intensive, especially for slides containing a low number of spermatozoa. Standard methods to identify spermatozoa are based on non-specific staining techniques such as hematoxylin/eosin (HE) and nuclear fast red/picroindigocarmine (CTS or “Christmas tree stain”) , which rely on sperm morphology for identification  and cannot be used for automated sperm detection. Moreover HE staining reduces the yield of DNA [4, 5, 11] and CTS staining results in DNA damage, probably due to the picric acid component .
Sperm HY-LITER™ (Independent Forensics, Hillside, IL, USA) is a fluorescent kit for the detection of human spermatozoa which does not rely on morphological characteristics or non-specific staining for identification. All relevant information concerning this kit can be found at http://www.spermhy-liter.com/. The spermatozoa are detected using an Alexa Fluor 488 derivatized mouse monoclonal antibody against proteins contained in the human sperm heads. In addition, a 4′,6-diamidino-2-phenylindole staining is used to detect all nuclei present on the slide. Processed slides can be viewed at low magnification, greatly increasing the speed of sperm identification.
The PALM MicroBeam system (P.A.L.M. Microlaser Technologies, Bernried, Germany), allows scanning of microscope slides . The system can be supplemented with image analyzing software modules allowing automated specimen identification and image processing  e.g. the AxioVision Commander (Carl Zeiss, Hallbergmoos, Germany). This image analyzing software can be optimized to automatically detect cells with specific features. In the current report we have optimized for the automatic detection of spermatozoa, stained with Sperm HY-LITER™ (Independent Forensics). The detected spermatozoa can then be collected by a defined laser pulse using the PALM laser pressure catapulting (LPC) function. This contact-free cell collection is an ideal approach to avoid sample contamination in forensic cases.
In the present study, an automated scanning method was developed to identify sperm heads, stained with Sperm HY-LITER™ (Independent Forensics), on smear preparations from postcoital vaginal swabs. LPC was used to catapult the auto-detected spermatozoa and DNA analysis was performed. After optimization of the DNA extraction protocol, the effect of the Sperm HY-LITER™ staining on downstream analysis of laser pressure catapulted spermatozoa was examined and the ability to obtain DNA profiles from spermatozoa catapulted from stained postcoital vaginal smears was assessed.
Materials and methods
Vaginal epithelial cells from healthy volunteers were collected onto sterile cotton swabs. These swabs were agitated in phosphate buffered saline (PBS, Gibco, Paisly, UK). After cell counting, PBS was added to obtain an epithelial cell working solution of 2 × 106 cells/ml. Semen samples were obtained from healthy volunteers. Semen working solutions of 2 × 106 spermatozoa/ml were prepared by diluting the liquid semen samples in PBS. Mixtures were prepared by combining semen working solution with epithelial cell working solution in the following ratios: 1:2, 1:10, and 1:50. Of each mixture 30 μl was used to make smear preparations on routine glass object slides (Dako silanized slides, Glostrup, Denmark). In addition, pure semen smears and postcoital vaginal swab smears were made.
The preparations were all dried overnight at room temperature and stained using the Sperm HY-LITER™ kit (Independent Forensics) according to manufacturer’s suggested protocol with slight modifications: the preparations were fixed for 2 min using two drops of a 70% ethanol solution (absolute ethanol, Merck BV, Schiphol-Rijk, The Netherlands) in pure water (MilliQ, Millipore, Billerica, MA, USA) in stead of the Sperm HY-LITER™ fixation solution. After the last washing step in the protocol, the slides were washed with 2 ml of pure water (MilliQ, Millipore) to remove adherent salts from the Sperm HY-LITER™ wash buffer and the stained slides were visualized immediately without mounting.
The scanning stage was controlled by the PALM RoboSoftware version 4 (P.A.L.M. Microlaser Technologies). Image acquisition was carried out with the AxioVision multichannel fluorescence module (Carl Zeiss) and the AxioCam MRm camera (Carl Zeiss).
The cell nuclei, including epithelial and sperm cell nuclei, were visualized using Zeiss filter set no. 49 (G 365 nm, FT 495, BP 445/50). The spermatozoa were visualized using Zeiss filter set no. 38 (BP 470/40, FT 495, BP 525/50). The slides were scanned at 20× magnification using a Carl Zeiss long distance Plan-Neofluar® objective. From every slide, 100 images were acquired using the scanning mode. The acquired images were displayed as an overview image in the PALM Navigator window and the individual images were stored as tiff-files.
Segmentation and masking
For automated detection of spermatozoa, the image processing AxioVision Commander module (Carl Zeiss) was used. All steps of processing, analysis, and evaluation were stored in an AxioVision Commander Script, which could be run automatically on the stored images.
In a first step the sperm heads, detected in the green fluorescence channel, were discriminated from the background using an interactive threshold. The resulting image was a binary image where the background was black and the detected regions were white. In a following step all detected regions below 3 μm or above 7 μm were removed from the binary image to eliminate artefacts smaller or bigger than a sperm head (approximately 5 μm).
Then the nuclei detected in the blue fluorescence channel were distinguished from the background using an interactive threshold. Afterwards artefacts (regions with an area above 70 μm2) were also eliminated from the resulting binary image. Following this procedure all nuclei still touching each other were separated using a watershed algorithm, to make sure every nucleus was counted separately. This algorithm splits the image into disjoint regions containing only one nucleus and is based on the topology of the image.
The coordinates of the detected spermatozoa were transferred automatically to an element list in the PALM RoboSoftware and after visual verification of the detected spermatozoa a catapulting point was set on the sperm head.
Laser pressure catapulting
The detected spermatozoa were collected by laser pressure catapulting using a pulsed nitrogen UV-A laser (wavelength 355 nm). The high energy generated by the focused laser light was used to catapult the detected spermatozoa into the cap of a standard 0.2 ml microfuge tube (Westburg, Leusden, The Netherlands) containing 18 μl of PicoPure DNA extraction buffer (PicoPure DNA extraction kit, Arcturus, Mountain View, CA, USA) supplemented with 2 μl of 1 M DTT (MP Biomedicals, Solon, OH, USA). This contact-free method avoids contamination of the sample.
DNA was extracted from the catapulted spermatozoa, using the PicoPure DNA extraction kit (Arcturus). The samples were incubated at 65°C overnight, centrifuged briefly and heated to 95°C for 30 min to inactivate proteinase K. If the polymerase chain reaction (PCR) could not be performed immediately, the samples were stored at −20°C.
Amplification and detection
For assessing the profile recovery after LPC from pure semen samples, a multiplex of four STR loci (CD4, TH01, D21S11, and SE33) was used as described earlier  with slight modifications. In short, 1.3 units of hotstar Taq DNA polymerase (Qiagen, Huntsville, AL, USA) was used and the samples were amplified on an Applied Biosystems GeneAmp 9700 60 Well thermal cycler (Foster City, CA, USA). Amplification parameters were: preincubation at 95°C for 15 min, followed by 33 cycles of denaturation for 60 s at 94°C, annealing for 60 s at 58°C and extension for 80 s at 72°C. This was followed by a final elongation step of 5 min at 72°C. At the end of the PCR reaction the temperature was kept at 4°C.
For the amplification of the DNA extracted from the spermatozoa isolated by LPC from the postcoital samples, the AmpFℓSTR® Profiler Plus® kit from Applied Biosystems was used according to the manufacturer's instructions, except that Hotstar Taq DNA polymerase was used instead of Taq Gold. The total number of cycles was 33.
The amplified fragments were then separated and analyzed by capillary electrophoresis using an ABI 3100 Genetic Analyzer (Applied Biosystems).
Automated detection of spermatozoa
As the Sperm HY-LITER™ kit (Independent Forensics) has been designed for microscopic screening of mounted slides, a few modifications of the manufacturer’s suggested protocol had to be carried out to make it compatible with LPC. An additional washing step with pure water was added to the protocol, to remove salt crystals originating from the Sperm HY-LITER™ wash buffer. This was necessary because the salt crystals were highly autofluorescent in the green fluorescence channel and interfered with the detection of the stained sperm heads. After this additional washing step, the slides could be visualized unmounted, which is necessary for LPC.
Automatically detected spermatozoa and nuclei in 100 acquired images per slide
Ratio of spermatozoa/nuclei used to make the smears
Number of sperm heads detected
Total number of nuclei (spermatozoa and epithelial cells) detected
Detected ratio of spermatozoa/nuclei
The postcoital vaginal swab smear was re-examined manually and only 2.8% false positives were found. Both the low amount of false positives and the good correlation between the ratio used to make the smears and the detected ratio, show the very high reliability of the developed AxioVision Commander Script.
The detected spermatozoa were catapulted by LPC in to a standard microfuge tube. Images acquired before and after catapulting clearly show that the catapulting process removes the sperm head from the glass slide.
Profile recovery after laser pressure catapulting
Three DNA isolation methods were evaluated (data not shown): an alkaline extraction method , DNA IQ™ System (Promega Corporation, Madison, WI, USA) and PicoPure DNA extraction kit (Arcturus) and the best results were obtained with the latter. After optimization, this DNA extraction method was used for further assessment of profile recovery after LPC.
Different amounts of spermatozoa (200, 100, 75, 50, 40, 30, 20, and 10) were collected by LPC from the pure sperm smears to assess the effect of the Sperm HY-LITER™ staining on the recovery of DNA. A full DNA profile of the four assessed loci (CD4, TH01, D21S11, and SE33) could be recovered from every sample. This proves that the Sperm HY-LITER™ staining has no negative influence on the DNA quality of the spermatozoa.
Profile recovery from spermatozoa after LPC from postcoital samples
Number of isolated spermatozoa
Number of expected male alleles
Number of detected male alleles
Number of contaminating female alleles
The results demonstrate that the AxioVision Commander Script has sufficient sensitivity to identify sperm heads stained with Sperm HY-LITER™ (Independent Forensics), present in an overwhelming background of epithelial cells. The sperm heads and the cell nuclei are easily detectable and the background staining is negligible. The nucleus staining provides additional confirmation that the Alexa Fluor 488 signal observed is related to a DNA-containing sperm head. The coordinates of the single cells assessed as spermatozoa are transferred to an element list of the PALM RoboSoftware. Before laser catapulting, the detected cells can be verified easily through bright field microscopy based on sperm cell morphology.
The scanning, analysis and identification of spermatozoa can be performed on normal microscopic slides, which are used with the Belgian sexual assault sets, allowing its use in existing preparations from sexual assault cases. Processed slides can be scanned at low magnification, greatly increasing the speed of sperm identification.
For forensic DNA profiling, it is important that the staining reagents used have no adverse effects on the recovery of DNA from spermatozoa on postcoital vaginal swabs. As robust DNA profiles without allelic drop out could be obtained from as little as 30 spermatozoa recovered from postcoital samples after LPC, it can be concluded that the Sperm HY-LITER™ staining has no influence on downstream DNA analysis, in contrast to traditional non-specific sperm-staining methods as HE and CTS.
In one of the postcoital samples, female DNA was present in the profile after LPC. This may be due to adherence of female DNA to the spermatozoa [7, 16]. Despite the presence of female DNA in one of the samples after LPC, the male DNA profile was interpreted easily. Nevertheless, further work will concentrate on generating pure male DNA profiles from this kind of samples.
As the whole procedure of staining, scanning, and catapulting can be performed in half a day, it can be concluded that, used in combination with the automatic detection of spermatozoa by the AxioVision Commander Script, the LCM method is a fast, sensitive and non-contact procedure for DNA profiling in cases of sexual assault.
The Research was funded by a PhD grant from the Institute for the Promotion of Innovation through Science and Technology in Flanders (IWT-Vlaanderen) to Mado Vandewoestyne.
The authors are grateful to Wouter Christiaens (Zeiss, Belgium) and Renate Burgemeister & colleagues (P.A.L.M. Microlaser, Germany) for their help in the development of the AxioVision Commander Script.
The authors declare that they have no competing financial, personal nor professional interests.
This article is distributed under the terms of the Creative Commons Attribution Noncommercial License which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.