Consequences of digital mammography in population-based breast cancer screening: initial changes and long-term impact on referral rates
- First Online:
- Cite this article as:
- Bluekens, A.M.J., Karssemeijer, N., Beijerinck, D. et al. Eur Radiol (2010) 20: 2067. doi:10.1007/s00330-010-1786-7
- 754 Downloads
To investigate the referral pattern after the transition to full-field digital mammography (FFDM) in a population-based breast cancer screening programme.
Preceding the nationwide digitalisation of the Dutch screening programme, an FFDM feasibility study was conducted. Detection and referral rates for FFDM and screen-film mammography (SFM) were compared for first and subsequent screens. Furthermore, radiological characteristics of referrals in digital screening were assessed.
A total of 312,414 screening mammograms were performed (43,913 digital and 268,501 conventional), with 4,473 consecutive referrals (966 following FFDM). Initially the FFDM referral rate peaked, and many false-positive results were noted as a consequence of pseudolesions and increased detection of (benign) microcalcifications. A higher overall referral rate was observed in FFDM screening in both first and subsequent examinations (p < .001), with a significant increase in cancer detection (p = .010).
As a result of initial inexperience with digital screening images implementing FFDM in a population-based breast cancer screening programme may lead to a strong, but temporary increase in referral. Dedicated training in digital screening for radiographers and screening radiologists is therefore recommended. Referral rates decrease and stabilise (learning curve effect) at a higher level than in conventional screening, yet with significantly enhanced cancer detection.
KeywordsBreast cancer Digital mammography Mass screening Referral rate Training
Screening mammography has been shown to be effective in reducing breast cancer mortality [1, 2]. Full-field digital mammography (FFDM) is increasingly used in the clinical setting, with a number of advantages resulting in better detection of (early) breast cancer and improvements in workflow .
Because of innovations, fast technical developments and emerging evidence that FFDM as a screening test is at least as effective as conventional screen-film mammography (SFM), the national coordinating team of the Dutch breast cancer screening decided to start digital pilot projects in 2003. Their main aim was to gain experience and to test the feasibility of digitalisation of the nationwide Dutch screening programme.
More recent studies concerning diagnostic accuracy confirmed that digital mammography as a screening technique in a population-based screening programme is at least as accurate as SFM [4, 5, 6, 7, 8, 9, 10, 11]. In several studies an increase in referral rate following the implementation of FFDM has been reported [4, 5, 8], a tendency we also recognised in our previous study . Nevertheless, this effect on referral rate has never been analysed in detail. Since changes in referral rate directly affect the daily screening practice and work-up, understanding this increase in referral rate is of crucial importance, and analysis could be useful to design a dedicated training programme.
The focus of this study, as part of the Dutch digital pilot projects, was to investigate the referral pattern after the introduction of digital mammography in the national screening programme. We provide a comprehensive evaluation of the changes in referral rates over time, reported separately for first and subsequent screening examinations compared with those in conventional mammography. Additionally, a detailed analysis of the radiographic characteristics of the referred lesions will be given.
Materials and methods
The current observational study encompasses the results of the first 4 years of the digital mammography pilot, which started in October 2003 at Preventicon screening centre (Utrecht, the Netherlands). At that time digital mammography was introduced by replacing one of the conventional units by an FFDM system. The pilot was part of the national screening programme, which offers mammography to all women aged 50 to 75 years every 2 years. Participation is on the basis of a written invitation by mail based on information provided by the national population registry. Details concerning the national screening programme have been published elsewhere . When further assessment is indicated, women are referred to a breast team of a nearby hospital.
All women participating in the screening programme at Preventicon screening centre between October 2003 and December 2007 were included in the study. In the first year after the introduction of the FFDM system only women attending their first screening examination were offered digital mammography. One year after the introduction, FFDM also became available for women attending subsequent screens. Assignment of women to FFDM or SFM was based on the random availability of the units when participants presented at the screening centre. However, women who already had a previous digital screening mammogram were always offered FFDM.
Participants were informed in writing about the possibility of having a digital mammogram. They had the right to refuse this offer and undergo conventional mammography. To comply with privacy regulations, participants signed a general informed consent form which permits use of data from the screening programme for evaluation and scientific research. Refusal of sharing data for these purposes was extremely rare. The study was approved by the regional medical ethics review board. Specific written informed consent for this study was not required.
Image acquisition and interpretation
Screen-film mammograms were acquired on two types of systems (GE 600/800T, GE Healthcare, Buc, France). All digital mammograms were acquired using a Lorad Selenia FFDM system (Hologic Inc., Danbury, CT) with a 70-µm pixel size and a 232 × 286-mm field of view. First screening examinations always included mediolateral-oblique (MLO) and craniocaudal (CC) views. At subsequent screening examinations MLO views of each breast were routinely acquired and, when indicated, CC views were also obtained. Radiographers are extensively trained to decide on obtaining CC views in subsequent screening examinations (FFDM as well as SFM) based on prespecified criteria for breast density and visible abnormality. In the case of a subsequent screening examination (digitised) prior mammograms were available at all times. Computer aided detection (CAD) was freely available for FFDM (ImageChecker, Hologic/R2, Santa Clara, CA). Digital mammograms were interpreted using soft-copy reading. The same team of radiologists was involved in the double-reading of FFDM and SFM, with differences of opinion resolved by consensus. One of two radiologists (D.B., J.D.), both with more than 15 years of experience in mammography screening, was involved in each screening examination performed during the entire study period. They were complemented by a team of six screening radiologists, each performing over 5,000 readings per year. They all had at least 2 years’ experience with working in a digital radiology environment before the study started. None of the readers had experience with the use of FFDM in screening.
More detailed descriptions of image acquisition and interpretation, reading conditions, and diagnostic work-up used in the Preventicon pilot are published elsewhere .
Retrospectively we collected screening outcomes and demographic features of all women included in the study. Normal findings at assessment or benign breast disease (including lobular carcinoma in situ) were reported as a negative (or false positive) screening outcome, whereas a positive screening outcome indicated malignancy (including ductal carcinoma in situ). Radiological characteristics of all referred women screened digitally during the study period were reviewed.
Study data were collected from the Preventicon archive and digital database, which include number of women screened and referred, radiological reports and final diagnoses. A search of the nationwide network and archive of histocytopathology (PALGA) completed the data.
Abnormalities seen on mammograms followed by referral were categorised as: mass, calcifications and other.
All performance indicators were computed separately for first and subsequent screening examinations. Screening intervals were calculated for subsequent screening examinations by taking the period between the current and previous screen. Because screening intervals where somewhat different in the two populations we computed detection per 24 months, by multiplying the observed rates by 24/T, with T denoting the median screening interval. The difference occurred due to different logistics in the permanent facility where FFDM was installed, whereas most SFM units were functional in mobile facilities.
We compared referral rate, breast cancer detection and positive predictive value (PPV) for FFDM and SFM separately for first and subsequent screening examinations. In addition, radiological characteristics of referrals in digital screening were assessed.
Statistical software was used for data management and analysis (SPSS, version 15.0 for Windows; SPSS Inc., Chicago, IL, USA). Results were compared using the Chi-square test. A p value of less than 0.05 was considered statistically significant. Dot graphs were drawn with a Loess fitted curve to evaluate trends in referral rates for both modalities. Bar graphs were used to present changes in radiological characteristics of referred lesions and the concurrent screening results. For all graphs variables were assessed per 2 months.
During the 4-year study period a total of 312,415 screens were performed, consisting of 43,914 digital and 268,501 conventional mammography procedures. In the FFDM group a total of 966 women were referred, in the SFM group 3507. The mean age of referred women was 58.5 years for conventional screening and 57.4 years for digital screening (range 50–75 years). The median screening interval was 22.7 months in the group of referrals screened with FFDM and 24.6 months for referrals after SFM screening. Breast cancer was detected in 1,619 women of whom 250 had digital mammograms.
Screening performance of SFM compared with FFDM, for first and subsequent individual screening examinations
Screened women (n)
First screening examination
Subsequent screening examination
Referred womena (n, rate)
First screening examination
Subsequent screening examination
Breast cancers detected (n, rate)
First screening examination
Subsequent screening examination
PPV of referral (%)
First screening examination
Subsequent screening examination
Trends in referral and detection rate in conventional and digital screening
Within a few months the rate started to decline in both groups. Referral stabilised after a little more than 1 year of digital screening, however at a higher level than in conventional screening.
The high referral rate was reflected in the detection. In the first year the mean detection rate in digital screening was 7.8 per 1,000 women screened. Detection stabilised with decreasing referral rate, but overall remained higher than the detection in conventional screening: 5.9‰ versus 5.0‰, respectively. Referral and detection rates with SFM remain relatively constant over the entire study period.
Trends in digitally referred lesions
In accordance with previous European studies [4, 5, 6, 7, 8, 9, 10] we found an improved detection with digital mammography, although this time significantly. In first screening examinations, i.e. the youngest women of the screening population, the relative increase in detection was most prominent. This finding is in line with the results of the DMIST trial that first showed that FFDM performed markedly better in younger women than SFM . In our study, the overall referral rate was higher in digital screening mammography. Referral rates may fluctuate in time, and an increase may sometimes be observed, for example after a course or meeting is attended. Since the referral rate in conventional mammography did not change during that same period, such a temporary trend is highly unlikely. Therefore, the high referral rate may be ascribed to the introduction of digital screening. The increase in referral was most distinct in the first period when only first screens were performed. However, a following substantial increase was present when subsequent examinations were introduced while a part of the learning curve had already been passed. The increase in referral in turn led to a lower PPV of referral.
An increase in referral in FFDM has been reported by previous studies evaluating screening programmes resembling the Dutch one [4, 5, 8]. On the other hand, two studies describing digital screening mammography in Scandinavia, found the highest referral rate in SFM [6, 7]. Both Scandinavian studies involved a subset of the screening population, concerning subsequent screens or first screens only. Furthermore, the study by Heddson et al.  was performed after a 6-month learning period, which might have lowered the overall referral rate. Two studies conducted in Spain and the UK reported no differences in referral after transition to FFDM in their screening programme [9, 10]. The relatively high baseline referral rate in Spain (11.5% in first screens, and 3.6% in subsequent screens) and the UK (overall 4.4%) might explain this outcome.
While former studies [4, 5, 6, 7, 8, 9, 10, 11] focus on the aggregate screening results of FFDM compared with SFM, this study, to our knowledge, is the first to analyse the effects over time after the transition to digital screening mammography. Referral peaks were seen in the first months, followed by a rapid decrease. The overall referral rate stabilised at a higher level than in screen-film mammography. The significant rise in referral might have affected the cancer detection rate as the overall detection rate tends to be higher in FFDM. Our previous study showed this to be largely attributable to relatively more detected cancers in first screens .
Secondly, higher contrast resolution improves the detection of microcalcifications . This explains the increased share of referrals on the basis of microcalcifications and is in line with the significant increased detection of ductal carcinoma (both in situ and invasive) found in our previous study . Since most clusters of microcalcifications do not represent malignancy, interpreting calcification type is another challenge in FFDM. Therefore, it is not surprising that in the first few months of digital screening we have seen an exceedingly high proportion of false-positive referrals due to microcalcifications.
The superior contrast resolution of digital mammography had yet a third effect. Compared with a previous conventional image, a digital mammogram can falsely give the impression that a lesion or cluster of microcalcifications is new or has grown. This doubt about progression in the presence of conventional prior mammograms can cause more women to be referred. While in other studies prior mammograms were offered only on request  or not until consensus meeting , we had (digitised) prior mammograms present at all times. Prior mammograms increase the specificity of screening  and not presenting them instantaneously was considered unethical. Accordingly, the introduction of subsequent screens a few months after the pilot started resulted in a second rise in referral.
Thus, while the higher overall referral rate can be justified by the improved cancer detection, the peak in referral and the high proportion of false-positive results just after implementation of FFDM primarily reflected inexperience with the interpretation of digital images and the comparison with digitised conventional prior mammograms in the screening setting.
The design of our study included the use of CAD, because we believe that with digital mammography this will become standard practice in screening programs. The fact that previous studies without CAD also reported increased detection of microcalcifications suggests that this result should not be attributed to CAD alone. On the other hand, CAD tends to increase referral rate as well . However, this should not have influenced the course of referral over time.
Measurement of the effect of CAD as a separate variable was not a subject of this study. Most reports in the literature demonstrate a benefit of CAD when single reading is practised. However, double reading with consensus may yield comparable results . Our study shows a combined effect of double reading and CAD. Double reading minimises perception errors and improves decisions, but most likely reduces the incremental benefit of CAD. We therefore reckon the possible confounding effect of CAD in this study to be marginal.
Two weaknesses of this study need to be addressed. This study was performed on one type of mammography unit (i.e. Lorad Selenia) with its proprietary image processing and display. It is, therefore, not clear whether these results can be generalised to other manufacturers as well. A comparative multi-centre study is needed to evaluate the applicability of these results in all mammography units.
Secondly, a pilot setting in which screening radiologists read conventional as well as digital mammograms, may not be entirely representative for the screening programme setting in which all mammograms have the same imaging characteristics. The mixture of conventional and digital mammograms in the pilot setting may have contributed to the increase in referral.
Screening programmes around the world vary in many different aspects. For instance, compared with screening on the basis of self-referral as performed in the USA, population-based screening programmes are characterised by a relatively low baseline referral rate. The Netherlands in particular is known for its low referral rate. Generalisation of our findings to other screening programmes is therefore not straightforward. However, our study results might still be of interest to those programmes with a similar set-up, planning a transition to digital mammography.
In the Netherlands, as a consequence of our findings we developed a dedicated training in digital screening mammography for radiographers and screening radiologists. Preliminary reports suggest this training is effective in the Netherlands since a similar effect in newly digitalised screening centres is not observed. We therefore recommend dedicated training in digital mass screening when converting to FFDM in a population-based breast cancer screening programme, especially when characterised by a relatively low referral rate.
We like to thank Inge van Veggel for her help in collection and management of data.
This work was presented at the ECR 2010.
This article is distributed under the terms of the Creative Commons Attribution Noncommercial License which permits any noncommercial use, distribution, and reproduction in any medium, provided the original author(s) and source are credited.