Feasibility study of computed tomography colonography using limited bowel preparation at normal and low-dose levels study
- First Online:
- Cite this article as:
- Florie, J., van Gelder, R.E., Schutter, M.P. et al. Eur Radiol (2007) 17: 3112. doi:10.1007/s00330-007-0668-0
- 282 Views
The purpose was to evaluate low-dose CT colonography without cathartic cleansing in terms of image quality, polyp visualization and patient acceptance. Sixty-one patients scheduled for colonoscopy started a low-fiber diet, lactulose and amidotrizoic-acid for fecal tagging 2 days prior to the CT scan (standard dose, 5.8–8.2 mSv). The original raw data of 51 patients were modified and reconstructed at simulated 2.3 and 0.7 mSv levels. Two observers evaluated the standard dose scan regarding image quality and polyps. A third evaluated the presence of polyps at all three mSv levels in a blinded prospective way. All observers were blinded to the reference standard: colonoscopy. At three times patients were given questionnaires relating to their experiences and preference. Image quality was sufficient in all patients, but significantly lower in the cecum, sigmoid and rectum. The two observers correctly identified respectively 10/15 (67%) and 9/15 (60%) polyps ≥10 mm, with 5 and 8 false-positive lesions (standard dose scan). Dose reduction down to 0.7 mSv was not associated with significant changes in diagnostic value (polyps ≥10 mm). Eighty percent of patients preferred CT colonography and 13% preferred colonoscopy (P<0.001). CT colonography without cleansing is preferred to colonoscopy and shows sufficient image quality and moderate sensitivity, without impaired diagnostic value at dose-levels as low as 0.7 mSv.
KeywordsCT colonographyFecal taggingColonic neoplasmColonoscopy
computed tomography colonography
Since computed tomography colonography (CTC) has shown good results in detecting clinically relevant polyps [1–6], studies have been performed focusing on increasing patient acceptance by reducing the bowel preparation. This can be done by adding an oral contrast agent to meals (fecal tagging) [7–12] in combination with a low-fiber diet, and sometimes lactulose for stool softening, thereby obviating extensive cleansing.
Different studies have been performed using barium, iodine or a combination of both as tagging material. However, only a limited number of these studies on CTC have used limited bowel preparation. Few have evaluated the diagnostic value and patient acceptance of CTC. In parallel with the development of limited bowel preparation strategies, (ultra) low-dose protocols have been evaluated [13–15].
To our knowledge, to date no studies have evaluated the effect of dose reduction on sensitivity and specificity in patients with a limited bowel preparation. Only one study  has evaluated limited bowel preparation at a relatively low dose level (140 kVp, 10 mAs). This study has shown very good results; however, these data have not been reproduced yet. Moreover, as only one dose-level was studied, the effect of dose reduction remains unknown.
The purpose of this study was to evaluate limited bowel-preparation CTC using an oral contrast agent (amidotrizoic acid) in terms of image quality, patient acceptance and polyp visualization using conventional colonoscopy (CC) as a reference standard. A second objective was to determine the effect of substantially reducing the radiation dose levels on the diagnostic accuracy of limited bowel preparation CTC, again using CC as the reference standard.
Materials and methods
Patients of the Onze Lieve Vrouwe Gasthuis at increased risk for colorectal cancer (personal or family history of colorectal polyps or cancer)  who were scheduled to undergo CC between April 2002 and August 2003 were invited to participate in the study. Exclusion criteria were: impossibility to understand the information/informed consent, age below 18 years, pregnancy and inflammatory bowel disease. The study was approved by the institutional review board of the Onze Lieve Vrouwe Gasthuis and Academic Medical Center. All patients gave written informed consent.
CTC bowel preparation and scanning protocol
Four weeks prior to the CC patients were asked to ingest amidotrizoic acid (20 mg/ml, made by the hospital pharmacy, 11.7 mg I/ml; corresponding to approximately 30 times diluted Gastrografin370®) three times a day (100 ml at breakfast and lunch, 300 ml at dinner) with a low-fiber diet (well cooked vegetables and meat, no fibrous fruit, no whole-wheat products, no nuts) starting 2 days prior to CTC. Lactulose (12 g, lactulose CF powder 6 g/sachet, Centrafarm, Etten-Leur, The Netherlands) was taken in the morning for 3 days prior to CTC for stool softening. Twenty mg of butylscopolaminebromide (Buscopan; Boehringer-Ingelheim, Ingelheim, Germany), when contraindicated, 1 mg of glucagon hydrochloride (Glucagen; Novo-Nordisk, Bagsvaerd, Denmark) was administered intravenously. The colon was insufflated with a CO2-air mixture (13.2% vol.) using a flexible catheter until patients experienced discomfort (±2–3 l). Patients were scanned in prone and supine position with a four-slice MX8000 (Philips Medical Systems, Best, The Netherlands) CT scanner (120 kV, rotation time 0.75 s, pitch 1.25, collimation 4*2.5 mm, section thickness 3.2 mm, and reconstruction interval 1.6 mm, 50 or 70 milliampere-second (mAs); 70 mAs if the abdominal circumference was >102.5 cm, scan time 20–25 s).
Prior to CC, each patient was instructed to ingest 4 l of a macrogol solution (Colofort macrogol 4000 sachets, Ipsen, Hoofddorp, The Netherlands), starting on the evening prior to the CC. CC was performed by one of three experienced gastroenterologists of the Onze Lieve Vrouwe Gasthuis (experience as gastroenterologist: 23, 12 and 4 years) and recorded on videotape. Patients received sedatives [5 mg midazolam (Dormicum; Roche, Basel, Switzerland)] and analgesics [0.05 mg fentanyl (Janssen Pharmaceuticals, Beerse, Belgium)] at the request of the patient. If polyps were present, the location was determined and size was estimated, based on comparison with an open biopsy forceps prior to removal. A polyp was considered flat if its height was less than one-half of the diameter of the lesion . Only segments visualized at CC were used for analysis.
Simulation of low-dose CT colonography
To study polyp detection at lower dose, CTC examinations were simulated with a lower dose/mAs-value using an established simulation technique [18, 19] that has been applied in two earlier CTC studies [13, 14]. This simulation method consists of the controlled increase of noise in the raw CT transmission data prior to reconstruction of the images. In this way, scans were simulated at 20 mAs (in the order of 2.3 mSv for two scans: supine and prone) and 6 mAs (±0.7 mSv). A previous simulation study has demonstrated that the radiation dose of CTC after extensive bowel cleansing can be reduced down to 0.2 mSv . Since the contrast between tissue and tagged fecal material in limited bowel preparation colons is lower than that between air and tissue, we chose 0.7 mSv as the lowest simulated dose-level next to 2.3 mSv.
Scales used by observer 1 to rate image quality (upper part) and scales used by the patients to rate experience and preference
Image quality (patient, segment)*
1: poor, not diagnostic; 2: moderate, diagnostic with limitations; 3: good, diagnostic with minor limitations; 4: excellent, no limitations
1: collapsed; 2: poorly distended; 3: only moderately distended but segment is distended over its full length; 4: good; 5: very good
1: poor; 2: moderate; 3: good; 4: very good
Presence of stool (segmemt)
1: large amount of stool, segment fully filled; 2: moderate amount of stool, ~50% of lumen filled; 3: small amount of stool; 4: only contrast layer on the wall; 5: no stool at all
Most burdensome aspect
CTC preparation: diet, lactulose, contrast agent
CTC: iv puncture, catheter placing, insufflation, breathholds, prone position
CC: iv puncture, moving of scope, air insufflation, monitoring after CC
Not, mild, moderate, severe, extreme
Definitely CTC, probably CTC, possibly CTC, indifferent, possibly CC, probably CC, definitely CC
Most reluctant factor
CC, bowel preparation prior to the CC, CTC, the limited bowel preparation prior to the CTC
Polyp detection and image quality
A polyp detected at CTC was labeled as true positive if three criteria were met: segmental location and location within the segment corresponded with CC (when situated near the borders of the segment, localization in the adjacent segment was also accepted), the polyp size as estimated by the endoscopist (open forceps) corresponded with size as measured on CTC (50% margin based on the CC size was allowed), and appearance (morphology) closely resembled that of the corresponding polyp at videotaped CC. The unblinded research fellow determined the nature of false-positive findings ≥10 mm by reviewing the videotaped CC and CTC. If the polyp was possibly missed at CC, a repeat CC was called for.
Patient experience and preference were evaluated by six questionnaires (scales are shown in Table 1): 2 weeks prior to the CTC, directly after the CTC bowel preparation, after the CTC, after the CC preparation, after the CC and 5 weeks later. The first and the last questionnaires were sent by mail. After the CTC bowel preparation, patients were asked how burdensome they rated the CTC bowel preparation, and the most burdensome aspect. After CTC they rated how much pain they had experienced and how burdensome the CTC had been. Prior to CC patients were asked how burdensome the extensive bowel preparation had been. After the CC they were asked how much pain they had experienced and how burdensome the CC had been. After the CC and 5 weeks later, patients were asked for their preference for either CTC or CC assuming that both techniques were equally accurate, while in 20% of CTC examinations clinically relevant lesions would be shown, necessitating a therapeutic CC. Two weeks prior to the CTC, directly after the CC and 5 weeks later, patients were asked what they were most reluctant to undergo.
Differences in quality between CTC images of different segments were assessed using ordinal regression analysis. For this analysis we first determined the best segment and used this segment as reference segment. The regression coefficient of each segment estimates the change in the log transformed diagnostic odds ratio (DOR) compared to the reference segment. It can be interpreted, after antilogarithm transformation as relative diagnostic odds ratio (RDOR). A lower RDOR implies inferior image quality of the respective segment compared to the reference segment. Differences in polyp detection rates between CC and lower dose CTC were assessed with the McNemar test for paired proportions. The Wilcoxon signed rank test statistic was used to evaluate differences in patient experience between CTC and CC and to evaluate differences in preference between the questionnaire filled out directly after the procedures and 5 weeks later. The data were first dichotomized as preference for CTC versus preference for CC. These differences were tested using the chi-square test statistic. All P-values <0.05 were considered to indicate statistical significance.
Baseline characteristics of the study population
Age in years: mean ± sd† (range)
61 ± 12 (27–81)
H/O† colorectal polyps
H/O colorectal carcinoma
F/H† of colorectal polyps or cancer
Coexistent complains: Abdominal pain/hematochezia/altered bowel habits
Colonoscopy: number of polyps/patients with polyps:
≥10 mm initial colonoscopy
Morphology of polyps ≥6 mm (sessile/stalked/flat/CRC)
Morphology of polyps ≥10 mm (sessile/stalked/flat/CRC)
Colonoscopy: no. of patients receiving Sed+analg/sed/analg/none
Stool consistency prior to CTC (diarrhea/soft/normal)
Abdominal pain prior to CTC (major/minor)
Flatulence prior to CTC (major/minor)
Spasmolitycs during CTC (Buscopan/Glucagon/neither)
In five patients it was not possible to fully inspect the colon endoscopically. In one patient a 14-mm polyp was seen at CTC in a segment not inspected at CC and confirmed at surgery (not included in the analysis). In two patients, repeat CC showed a 30-mm polyp and a 30-mm carcinoma (included in the analyses), both missed at the initial CC.
Performance characteristics per observer per size category
Polyps ≥10 mm
Polyps ≥6 mm
Analysis according to polyp
Analysis according to patient
Polyp detection at lower dose-levels
Performance characteristics of observer 3 per dose level
Polyps ≥10 mm
Polyps ≥6 mm
Analysis according to polyp
4 /10 (40%)
Analysis according to patient
4 /9 (44%)
Patient experience and preference
This study demonstrates that CT colonography without cleansing is preferred to colonoscopy and shows moderate sensitivity (60–67%) for polyps ≥10 mm without impaired diagnostic value at mSv levels as low as 0.7 mSv.
Image quality was good on average. Nevertheless, the cecum, sigmoid and rectum showed overall reduced image quality. Although this can be caused by the pelvis causing more noise, reduced imaged quality must probably be attributed to inferior homogeneity in the cecum and rectum and inferior distension in the sigmoid and rectum in the prone position. In contrast to the known problems in CTC with distension, which are solved by dual positioning, inferior homogeneity in the cecum and rectum are typical for the limited bowel preparation protocol. Insufficient homogeneity in the cecum is probably caused by the fact that patients ate food after the last amount of contrast agent was taken, resulting in inadequately tagged stool. The inhomogeneous stool in the rectum probably was caused by the fact that in patients with a long transit time stool was already shaped before the fecal tagging was started.
We assume that two adaptations are necessary to improve image quality. First, contrast must be taken as long as the patients are eating. Second, stool should be made softer, for example by replacing lactulose by a stronger osmotic laxative (e.g., low-dose magnesium salts), thereby also reducing the transit time and reducing the amount of non-tagged stool in the rectum. Although these adaptations increase the burden of the bowel preparation, they are slight and are required to improve image quality and thereby maybe the sensitivity and specificity.
Our study showed an average sensitivity of 63% for large polyps (with a limited number of large false-positive lesions). This is lower than the 100% average sensitivity described by Iannaccone et al.  using a more or less comparable bowel preparation; no fiber-rich food and 200 ml of 370 mg/ml iodinated contrast agent in 48 h (which is 6.3 times the amount given in our study), but no lactulose. In addition to the differences in bowel preparation, differences in experience of the readers or reader performance, difference in study groups (polyp prevalence and spectrum) might explain these discrepant results. A recent study  has shown inferior results regarding diagnostic value of CTC in populations at increased risk, as is used in the current study, possibly due to a relatively high number of hard-to-see polyps. When comparing our study with three large studies on CTC with extensive bowel preparation [21–23] (showing per polyp sensitivities of respectively 52%, 32–73% and 53% for polyps ≥10 mm), our results are comparable or slightly better.
Since two of five (observer 1) and two of six (observer 2) missed polyps were not seen in retrospect by the unblinded observer, the problem of missing large polyps can be regarded as predominantly an interpretation problem rather than a visualization problem. Future developments, such as better homogeneity, better distension, better learning curve, intuitive display modes, electronic cleansing and computer-aided detection can help in reducing the number of missed polyps.
Sensitivity and specificity were impaired at lower dose levels for patients with lesions ≥10 mm. This is especially of interest in screening protocols. Regarding lesions ≥6 mm, specificity was significantly lower at the simulated 0.7 mSv low-dose scans. Increased noise levels made it impossible to see the tiny air bubbles inside stool. These results may seem to be in contrast with those in an earlier report on low radiation dose CTC  showing fewer false-positive lesions. However, these studies in patients with extensively cleansed colons used a three-dimensional display mode and filters to smooth the bowel wall. By using these smoothing filters, small artifacts and small amounts of residual stool were masked when evaluating the colon three dimensionally. Making stool softer and scanning with thinner slices possibly allows for better visualization of the air bubbles, thereby reducing the number of false positives.
Patient acceptance was found to be very good in the current study. Patient preference was convincingly in favor of CTC; 71% of patients preferred CTC with limited bowel preparation to CC for their next examination. Other studies have shown discrepant results on patient acceptance of CTC versus CC, some favoring CTC [24–27], others CC [28, 29]. In a study where patients first underwent extensive cleansing and consecutively CTC and CC  fewer patients (71% directly after the CC) favored CTC when compared to our study (80% directly after CC). This difference cannot be attributed to the bowel preparation because of a different study setup.
Several potential limitations must be considered. The number of patients included is relatively low, especially for determining the detection parameters at the 2.3 and 0.7 mSv levels.
In the current study no segmental unblinding during CC was performed. Although repeat CC showed that two large false-positive lesions were missed at the initial CC, segmental unblinding has the advantage of evaluating all false positives.
Since no electronic cleansing was used and many polyps were submerged, 3D evaluation of these lesions was not of additive value. However, 3D evaluation with electronic cleansing might have given better results, especially since most missed polyps were visible in retrospect.
Although the two readers that had no previous experience in CTC underwent a learning curve of 50 patients, some studies have shown that this may not be sufficient for optimal polyp detection [30, 31]. For CTC using limited bowel preparation, the learning curve is possibly even longer.
When filling out the questionnaires, sensitivity for CC and CTC were assumed equal. If patients knew that the sensitivity of CTC was lower than for CC, this most likely would have influenced the preference for CTC negatively.
In conclusion, these results show that CTC with the limited bowel preparation protocol used in this study is feasible, even when using doses as low as 0.7 mSv levels. Although sensitivity was not as high as previously reported, this is most likely not due to the limited bowel preparation, but to interpretation problems. Technical developments will probably increase the sensitivity, while minor adjustments in bowel preparation may reduce the number of false positives, especially when using low-dose protocols. Since patient acceptance was very good, this technique can be regarded as promising in screening patients of populations with a low prevalence of polyps.