High-quality recording of bioelectric events
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.Get Access
In the first part of the review the various mechanisms that could be the cause of interference in bioelectric recordings are considered. It is demonstrated that the performance of a good amplifier can be seriously degraded in its functioning if the whole measurement situation is not taken into account. Several techniques used to reduce interference, of which guarding and driven right leg circuits are the most important, are analysed. In the second part of the review some examples of the application of the theory in practical situations are presented. The instrumentation amplifier circuit normally used in bioelectric recordings is improved for measurements under difficult circumstances. Another application is a low-cost 64-channel amplifier for multichannel ECG recordings. The third application is a device that can be added to bioelectric measurement systems and will provide a major reduction in interference.
- Almasi, J. J. andSmith, O. H. (1970) Systemic and random variations of ECG electrode impedance.Ann. NY Acad. Sci.,170, 509–519.
- Forster, I. C. (1974) Measurement of patient body capacitance and a method of patient isolation in mains environments.Med. & Biol. Eng.,12, 730–732.
- Geddes, L. A. andBaker, L. E. (1966a) Chlorided silver electrodes.Med. Res. Eng.,6, 33–34.
- Geddes, L. A. andBaker, L. E. (1966b) The relationship between input impedance and electrode area in recording the ECG.Med. & Biol. Eng.,4, 439–450.
- Geddes, L. A. (1972)Electrodes and the measurements of bioelectric events. Wiley Interscience, New York, 44–94.
- Grimnes, S. (1983) Impedance measurements of individual skin surface electrodes.Med. & Biol. Eng. & Comput.,21, 750–755. CrossRef
- Hamstra, G. H., Peper, A. andGrimbergen, C. A. (1984) Low-power, low-noise instrumentation amplifier for physiological signals.,22, 272–274. CrossRef
- Huhta, J. C. andWebster, J. G. (1973) 60-Hz interference in electro-cardiography.IEEE Trans.,BME-20, 91–101.
- McClellan, A. D. (1981) Extracellular amplifier with boot-strapped input stage results in high common-mode rejection.Med. & Biol. Eng. & Comput.,19, 657–658. CrossRef
- Morrison, R. (1977)Grounding and shielding techniques in instrumentation, 2nd edn. John Wiley & Sons, New York, 132–133.
- Motchenbacher, C. D. andFitchen, F. C. (1972)Low-noise electronic design. John Wiley & Sons, New York, 185–187.
- Olson, W. H. (1978) Electrical safety. InMedical instrumentation: application and design.Webster, J. G. (Ed.), Houghton Mifflin Co., Boston, 667–706.
- Pacela, A. F. (1967) Collecting the body's signals.Electronics,40, 103–112.
- Pallas-Arény, R. (1988) Interference-rejection characteristics of biopotential amplifiers: a comparative analysis.IEEE Trans.,BME-35, 953–959.
- Peper, A., Jonges, R., Losekoot, T. G. andGrimbergen, C. A. (1985) Recording of surface His-Purkinje potentials.Med. & Biol. Eng. & Comput.,23, 365–376. CrossRef
- Reek, E. J., Grimbergen, C. A. andvan Oosterom, A. (1984) A low-cost 64 channel microcomputer based data acquisition system for bedside registration of body surface maps. Proc. 11th Int. Congr. Electrocardiol., Caen, France, 17th–20th July, 37–38.
- Smit, H. W., Verton, K. andGrimbergen, C. A. (1987) A low-cost multichannel preamplifier for physiological signals.IEEE Trans. BME-34, 307–310.
- Spekhorst, H., SippensGroenewegen, A., David, G. K., Metting van Rijn, A. C. andBroekhuysen, P. (1988) Radiotransparent carbon fibre electrode for ECG recordings in the catheterization laboratory.BME-35, 402–406.
- Thakor, N. V. andWebster, J. G. (1980) Ground-free ECG recording with two electrodes..BME-27, 699–704.
- Tobey, G. E., Graeme, J. G. andHuelsman, L. P. (1971)Operational amplifiers: design and applications. McGraw-Hill, New York, 206–207.
- Winter, B. B. andWebster, J. G. (1983) Driven-right-leg circuit design.IEEE Trans.,BME-30, 62–66.
- High-quality recording of bioelectric events
Medical and Biological Engineering and Computing
Volume 28, Issue 5 , pp 389-397
- Cover Date
- Print ISSN
- Online ISSN
- Kluwer Academic Publishers
- Additional Links
- Bioelectric recordings
- Driven right leg
- Industry Sectors