When are actively balanced biphasic (‘Lilly’) stimulating pulses necessary in a neurological prosthesis? I Historical background; Pt resting potential;Q studies
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.Get Access
Actively balanced (‘Lilly’) stimulating current waveforms are generally considered to give very ‘safe’ stimulation. Although this is perfectly true, the specification of the necessary waveform generators in neurological prostheses demands additional complexity, and probably additional expense and development time as well. The paper and its companion enquire whether the use of simple, passively charge-balanced stimulating pulses is equally safe, provided the stimulation parameters and circuitry are designed with appropriate care. It is concluded that, in respect of safe deliverable charge density per pulse at the electrode, release of noxious products and stimulating effectiveness, simple pulses need give no worse performance; in some circumstances they may give better.
- Berkley, C. (1965) Potential problems in long term stimulation. Digest of the 6th International Conference on Medical and Biological Engineering, Tokyo.
- Bockris, J. O'M. andReddy, A. K. N. (1970)Modern electrochemistry. Plenum Press, New York.
- Bowden, F. P. (1929) The amount of hydrogen and oxygen present on the surface of a metallic electrode.Proc. R. Soc.,125A, 446–462.
- Brummer, S. B. andTurner, M. J. (1977a) Electrochemical considerations for safe electrical stimulation of the nervous system with platinum electrodes.IEEE Trans.,BME-24, 59–62.
- Brummer, S. B. andTurner, M. J. (1977b) Electrical stimulation with Pt electrodes I.-—Ibid.,BME-24, 436–439.
- Brummer, S. B. andTurner, M. J. (1977c) Electrical stimulation with Pt electrodes II.-—Ibid.,BME-24, 440–443.
- Brummer, S. B., McHardy, J. andTurner, M. J. (1977) Electrical stimulation with Pt electrodes III.Brain Behav. Evol.,14, 10–22.
- de Boer, R. W. andvan Oosterom, A. (1978) Electrical properties of platinum electrodes: impedance measurements and time-domain analysis.Med. & Biol. Eng. & Comput.,16, 1–10.
- Donaldson, N. de N. andDonaldson, P. E. K. (1986) When are actively balanced biphasic (‘Lilly’) stimulating pulses necessary in neurological prostheses? II pH changes; noxious products; electrode corrosion; discussion.-—Ibid.,24, 50–56.
- Greatbatch, W. (1973) Electrochemical criteria for selection of chronically implanted stimulatory electrodes. InChronically implanted cardiovascular instrumentation.McCutcheon, P. (Ed.), Academic Press, New York, Chap. 19.
- Hoare, J. P. (1967)Advances in electrochemistry and electrochemical engineering, vol. 6 Delahay, P. (Ed.), Interscience, New York.
- Lilly, J. C., Austin, G. M. andChambers, W. W. (1952) Threshold movements produced by excitation of cerebral cortex and efferent fibres with some parametric regions of rectangular current pulses.J. Neurophysiol.,15, 319–341.
- Lilly, J. C., Hughes, J. R., Alvord, E. C. andGalkin, T. W. (1955) Brief non-injurious electric waveforms for stimulation of the brain.Science,121, 468–469.
- Loeb, J. C., McHardy, J., Kelliher, E. M. andBrummer, S. B. (1982) Neural prostheses. InBiocompatibility in clinical practice, Vol. II Williams, D. F. (Ed.), CRC Press. Chap. 8.
- McHardy, J., Robblee, L. S., Marston, J. M. andBrummer, S. B. (1980) Electrical stimulation with Pt electrodes IV.Biomaterials,1, 129–134. CrossRef
- Onaral, B. andSchwan, H. P. (1982) Linear and nonlinear properties of platinum electrode polarisation. Part 1: frequency dependence at very low frequencies.Med. & Biol. Eng. & Comput.,20, 299–306. CrossRef
- Peters, D. G. andLingane, J. J. (1962) Anodic formation and chemical analysis of oxychloride films on platinum electrodes.J. Electroanal. Chem.,4, 193–217. CrossRef
- Robblee, L. S., McHardy, J., Marston, J. M. andBrummer, S. B. (1980) Electrical stimulation with Pt electrodes B.Biomaterials,1, 135–139. CrossRef
- Robblee, L. S., Lefko, J. L. andBrummer, S. B. (1983) Activated Ir: An electrode suitable for reversible charge injection in saline solution.J. Electrochem. Soc.,130, 731–733. CrossRef
- Weinman, J. andMahler, J. (1964) An analysis of electrical properties of metal electrodes.Med. Electron. Biol. Eng.,2, 299–309.
- When are actively balanced biphasic (‘Lilly’) stimulating pulses necessary in a neurological prosthesis? I Historical background; Pt resting potential;Q studies
Medical and Biological Engineering and Computing
Volume 24, Issue 1 , pp 41-49
- Cover Date
- Print ISSN
- Online ISSN
- Kluwer Academic Publishers
- Additional Links
- Industry Sectors