Characterization and use of the new NIST rapid pneumatic tube irradiation facility
- D. A. Becker
- … show all 1 hide
Rent the article at a discountRent now
* Final gross prices may vary according to local VAT.Get Access
Recently a new rapid pneumatic tube facility was inserted into a long unused location in the NIST 20 MW nuclear reactor. This facility was designed and constructed specifically for rapid INAA using short lived activation products. Included is a computer controlled console which uses fast sensors to accurately measure the irradiation capsule flight time, and a loss-free counting system connected to a 32% efficient PHGe detector with a transistor reset preamplifier. Measurement of travel, times from end-of-irradiation to detector were 473±8 ms. Measurement of the thermal neutron fluence rate was 5.0·1013 n·cm−2·s−1. The other three pneumatic tubes in the NIST reactor have transfer times of 3 to 15 seconds, and no timing capability more accurate than human response. This new facility substantially improves our ability to accurately determine activation products with half-lives from 1 to 100 seconds. Characterization information reported on this new irradiation facility includes absolute fluence measurements, fluence rate variations within the capsule and variations with time, and determination of analytical sensitivities for fluorine-20 selenium-77m, and silver-110g.
- R. R. Greenberg, H. M. Kingston, R. L. Watters, Jr., K. W. Pratt, Fresenius J. Anal. Chem., 338, (1990) 394. CrossRef
- D. A. Becker, R. M. Lindstrom, J. K. Langland, R. R. Greenberg, J. Trace Microprobe Tech., 14 (1996), No. 1, 1.
- R. M. Lindstrom, D. A. Becker, J. K. Langland, R. R. Greenberg, J. Radioanal. Nucl. Chem., 215 (1997) 47. CrossRef
- D. A. Becker, NBS Tech. Note 548, U.S. Dept. Commerce, 6–23 (1970).
- D. A. Becker, P. D. LaFleur, J. Radioanal. Chem. 19 (1974) 149.
- D. A. Becker, J. Radioanal. Chem., 110 (1987) 393.
- W. N. McElroy, L. S. Kellogg, W. Y. Matsumoto, J. L. Jackson Hanford Engineering Development Laboratory, Monthly Progress Report, U.S. Atomic Energy Commission, Washington, DC, Oct. 1972, p. 1.
- Characterization and use of the new NIST rapid pneumatic tube irradiation facility
Journal of Radioanalytical and Nuclear Chemistry
Volume 233, Issue 1-2 , pp 155-160
- Cover Date
- Print ISSN
- Online ISSN
- Kluwer Academic Publishers
- Additional Links
- Industry Sectors
- D. A. Becker (1)
- Author Affiliations
- 1. Nuclear Methods Group, Analytical Chemistry Division/CSTL, National Institute of Standards and Technology, 20899-0001, Gaithersburg, MD, USA